-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180411_110123_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.000092 to 42.430620
-each separately from 0s to 42s and from 92us to 430620us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.010615 in 1 packets - #/No: 18/18
      27.419046 in 1 packets - #/No: 36/18
MIU packet time from 2018-04-11T11:01:13.000.092.000 to 2018-04-11T11:01:55.430.620.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.010615 in 1 packets - #/No: 18/18
      27.419046 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4393492 to 43.4391906
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999434 in 1 packets - #/No: 18/18
      26.9998980 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180411_110123_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4394419 to 1.4397309
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0105590  -- OK for SL --
   MIU time from 2018-04-11T11:01:13.000.092.000 to 2018-04-11T11:01:13.000.381.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010615

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4502899 to 16.4505789
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 27.4189430  -- OK for SL --
   MIU time from 2018-04-11T11:01:28.010.996.000 to 2018-04-11T11:01:28.011.285.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 27.419046

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.8695219 to 43.8698109
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-04-11T11:01:55.430.331.000 to 2018-04-11T11:01:55.430.620.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-04-11T11:01:12.999.951.750
   Max SL corrected packet time: 2018-04-11T11:01:55.430.479.750
   Min SL corrected XO rel time: 0.0000927
   Max SL corrected XO rel time: 42.4306219
   TM values from -13 to 5 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-04-11T11:01:12.999.951.750 (XO: 0.0000927, t=35.6°C)  Rank: 00001
   TM values: Max =     4  Min =   -13  Ave =    -4.44  Std =     2.49
   Min TM value:-13 at an offset of 49.813 usec -> 2018-04-11T11:01:13.000.001.562 (XO: 0.0001425)
   Max TM value:4 at an offset of 53.775 usec -> 2018-04-11T11:01:13.000.005.525 (XO: 0.0001465)
   Max dTM/dt [1/us]::6.40e+01 at an offset of 122.138 usec -> 2018-04-11T11:01:13.000.073.887 (XO: 0.0002148)
   Min dTM/dt [1/us]::-6.40e+01 at an offset of 122.563 usec -> 2018-04-11T11:01:13.000.074.312 (XO: 0.0002153)

 - Block number: 2 starting at 2018-04-11T11:01:28.010.855.750 (XO: 15.0109973, t=35.6°C)  Rank: 00001
   TM values: Max =     3  Min =   -12  Ave =    -4.41  Std =     2.29
   Min TM value:-12 at an offset of 9.838 usec -> 2018-04-11T11:01:28.010.865.587 (XO: 15.0110072)
   Max TM value:3 at an offset of 10.725 usec -> 2018-04-11T11:01:28.010.866.475 (XO: 15.0110081)
   Max dTM/dt [1/us]::4.80e+01 at an offset of 150.301 usec -> 2018-04-11T11:01:28.011.006.050 (XO: 15.0111476)
   Min dTM/dt [1/us]::-4.90e+01 at an offset of 184.301 usec -> 2018-04-11T11:01:28.011.040.050 (XO: 15.0111816)

 - Block number: 3 starting at 2018-04-11T11:01:55.430.190.750 (XO: 42.4303329, t=35.6°C)  Rank: 00002
   TM values: Max =     5  Min =   -12  Ave =    -4.51  Std =     2.25
   Min TM value:-12 at an offset of 25.838 usec -> 2018-04-11T11:01:55.430.216.587 (XO: 42.4303587)
   Max TM value:5 at an offset of 106.925 usec -> 2018-04-11T11:01:55.430.297.675 (XO: 42.4304398)
   Max dTM/dt [1/us]::6.60e+01 at an offset of 152.413 usec -> 2018-04-11T11:01:55.430.343.163 (XO: 42.4304853)
   Min dTM/dt [1/us]::-6.00e+01 at an offset of 143.913 usec -> 2018-04-11T11:01:55.430.334.663 (XO: 42.4304768)
