-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180407_221705_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 8.142692 to 30.862524
-each separately from 8s to 30s and from 142692us to 862524us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.259858 in 1 packets - #/No: 18/18
      5.459107 in 1 packets - #/No: 36/18
MIU packet time from 2018-04-07T22:17:02.241.692.000 to 2018-04-07T22:17:24.961.524.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.259858 in 1 packets - #/No: 18/18
      5.459107 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.4400510 to 31.4399663
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      16.9999346 in 1 packets - #/No: 18/18
      4.9999807 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      06 -> 47.2°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180407_221705_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.5827441 to 9.5830331
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.2597910  -- OK for SL --
   MIU time from 2018-04-07T22:17:02.241.692.000 to 2018-04-07T22:17:02.241.981.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.259858

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.8428241 to 26.8431131
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.4590870  -- OK for SL --
   MIU time from 2018-04-07T22:17:19.501.839.000 to 2018-04-07T22:17:19.502.128.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.459107

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.3022001 to 32.3024891
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-04-07T22:17:24.961.235.000 to 2018-04-07T22:17:24.961.524.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-04-07T22:17:02.241.551.750
   Max SL corrected packet time: 2018-04-07T22:17:24.961.383.750
   Min SL corrected XO rel time: 8.1426937
   Max SL corrected XO rel time: 30.8625261
   TM values from -16 to 6 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-04-07T22:17:02.241.551.750 (XO: 8.1426937, t=35.6°C)  Rank: 00002
   TM values: Max =     5  Min =   -14  Ave =    -4.53  Std =     2.66
   Min TM value:-14 at an offset of 223.488 usec -> 2018-04-07T22:17:02.241.775.238 (XO: 8.1429171)
   Max TM value:5 at an offset of 234.051 usec -> 2018-04-07T22:17:02.241.785.800 (XO: 8.1429277)
   Max dTM/dt [1/us]::1.11e+02 at an offset of 255.913 usec -> 2018-04-07T22:17:02.241.807.663 (XO: 8.1429496)
   Min dTM/dt [1/us]::-1.07e+02 at an offset of 34.088 usec -> 2018-04-07T22:17:02.241.585.837 (XO: 8.1427277)

 - Block number: 2 starting at 2018-04-07T22:17:19.501.698.750 (XO: 25.4028401, t=35.6°C)  Rank: 00002
   TM values: Max =     5  Min =   -15  Ave =    -4.51  Std =     2.57
   Min TM value:-15 at an offset of 252.063 usec -> 2018-04-07T22:17:19.501.950.813 (XO: 25.4030921)
   Max TM value:5 at an offset of 40.238 usec -> 2018-04-07T22:17:19.501.738.987 (XO: 25.4028803)
   Max dTM/dt [1/us]::9.00e+01 at an offset of 233.701 usec -> 2018-04-07T22:17:19.501.932.450 (XO: 25.4030738)
   Min dTM/dt [1/us]::-9.80e+01 at an offset of 50.413 usec -> 2018-04-07T22:17:19.501.749.162 (XO: 25.4028905)

 - Block number: 3 starting at 2018-04-07T22:17:24.961.094.750 (XO: 30.8622371, t=35.6°C)  Rank: 00002
   TM values: Max =     6  Min =   -16  Ave =    -4.45  Std =     3.00
   Min TM value:-16 at an offset of 67.713 usec -> 2018-04-07T22:17:24.961.162.462 (XO: 30.8623048)
   Max TM value:6 at an offset of 39.675 usec -> 2018-04-07T22:17:24.961.134.425 (XO: 30.8622768)
   Max dTM/dt [1/us]::1.22e+02 at an offset of 121.988 usec -> 2018-04-07T22:17:24.961.216.737 (XO: 30.8623591)
   Min dTM/dt [1/us]::-1.22e+02 at an offset of 112.125 usec -> 2018-04-07T22:17:24.961.206.875 (XO: 30.8623492)
