-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180404_110349_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 12.017324 to 30.690788
-each separately from 12s to 30s and from 17324us to 690788us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      10.625415 in 1 packets - #/No: 18/18
      8.047182 in 1 packets - #/No: 36/18
MIU packet time from 2018-04-04T11:03:49.116.324.000 to 2018-04-04T11:04:07.789.788.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      10.625415 in 1 packets - #/No: 18/18
      8.047182 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.4333259 to 31.4332573
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      9.9999619 in 1 packets - #/No: 18/18
      7.9999695 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      06 -> 47.2°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180404_110349_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.4506511 to 13.4509401
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.6253750  -- OK for SL --
   MIU time from 2018-04-04T11:03:49.116.324.000 to 2018-04-04T11:03:49.116.613.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.625415

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.0763151 to 24.0766041
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.0471510  -- OK for SL --
   MIU time from 2018-04-04T11:03:59.742.028.000 to 2018-04-04T11:03:59.742.317.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.047182

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.1237551 to 32.1240441
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-04-04T11:04:07.789.499.000 to 2018-04-04T11:04:07.789.788.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-04-04T11:03:49.116.183.750
   Max SL corrected packet time: 2018-04-04T11:04:07.789.647.750
   Min SL corrected XO rel time: 12.0173253
   Max SL corrected XO rel time: 30.6907894
   TM values from -13 to 4 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-04-04T11:03:49.116.183.750 (XO: 12.0173253, t=34.0°C)  Rank: 00002
   TM values: Max =     2  Min =   -11  Ave =    -4.56  Std =     1.71
   Min TM value:-11 at an offset of 17.575 usec -> 2018-04-04T11:03:49.116.201.325 (XO: 12.0173428)
   Max TM value:2 at an offset of 4.563 usec -> 2018-04-04T11:03:49.116.188.312 (XO: 12.0173298)
   Max dTM/dt [1/us]::5.00e+01 at an offset of 117.450 usec -> 2018-04-04T11:03:49.116.301.200 (XO: 12.0174427)
   Min dTM/dt [1/us]::-4.80e+01 at an offset of 103.763 usec -> 2018-04-04T11:03:49.116.287.512 (XO: 12.0174290)

 - Block number: 2 starting at 2018-04-04T11:03:59.741.887.750 (XO: 22.6430297, t=34.0°C)  Rank: 00002
   TM values: Max =     4  Min =   -13  Ave =    -4.48  Std =     2.74
   Min TM value:-13 at an offset of 115.700 usec -> 2018-04-04T11:03:59.742.003.450 (XO: 22.6431454)
   Max TM value:4 at an offset of 101.275 usec -> 2018-04-04T11:03:59.741.989.025 (XO: 22.6431310)
   Max dTM/dt [1/us]::6.00e+01 at an offset of 97.938 usec -> 2018-04-04T11:03:59.741.985.687 (XO: 22.6431277)
   Min dTM/dt [1/us]::-5.60e+01 at an offset of 101.488 usec -> 2018-04-04T11:03:59.741.989.237 (XO: 22.6431312)

 - Block number: 3 starting at 2018-04-04T11:04:07.789.358.750 (XO: 30.6905004, t=34.0°C)  Rank: 00002
   TM values: Max =     3  Min =   -11  Ave =    -4.40  Std =     1.99
   Min TM value:-11 at an offset of 35.250 usec -> 2018-04-04T11:04:07.789.394.000 (XO: 30.6905357)
   Max TM value:3 at an offset of 171.251 usec -> 2018-04-04T11:04:07.789.530.000 (XO: 30.6906717)
   Max dTM/dt [1/us]::5.80e+01 at an offset of 171.201 usec -> 2018-04-04T11:04:07.789.529.950 (XO: 30.6906716)
   Min dTM/dt [1/us]::-5.60e+01 at an offset of 10.388 usec -> 2018-04-04T11:04:07.789.369.137 (XO: 30.6905108)
