-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180402_194319_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 2.948791 to 40.283767
-each separately from 2s to 40s and from 283478us to 949080us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      21.722914 in 1 packets - #/No: 18/18
      15.611195 in 1 packets - #/No: 36/18
MIU packet time from 2018-04-02T19:43:10.948.791.000 to 2018-04-02T19:43:48.283.767.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      21.722914 in 1 packets - #/No: 18/18
      15.611195 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.4395021 to 41.4393567
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      21.9999159 in 1 packets - #/No: 18/18
      15.9999387 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180402_194319_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.3882913 to 4.3885803
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.7228310  -- OK for SL --
   MIU time from 2018-04-02T19:43:10.948.791.000 to 2018-04-02T19:43:10.949.080.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.722914

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.1114113 to 26.1117003
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.6111350  -- OK for SL --
   MIU time from 2018-04-02T19:43:32.671.994.000 to 2018-04-02T19:43:32.672.283.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.611195

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.7228353 to 41.7231243
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-04-02T19:43:48.283.478.000 to 2018-04-02T19:43:48.283.767.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-04-02T19:43:10.948.650.750
   Max SL corrected packet time: 2018-04-02T19:43:48.283.626.750
   Min SL corrected XO rel time: 2.9487928
   Max SL corrected XO rel time: 40.2837687
   TM values from -39 to 31 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-04-02T19:43:10.948.650.750 (XO: 2.9487928, t=35.6°C)  Rank: 00005
   TM values: Max =    23  Min =   -37  Ave =    -4.36  Std =     7.35
   Min TM value:-37 at an offset of 117.788 usec -> 2018-04-02T19:43:10.948.768.537 (XO: 2.9489106)
   Max TM value:23 at an offset of 53.400 usec -> 2018-04-02T19:43:10.948.704.150 (XO: 2.9488462)
   Max dTM/dt [1/us]::3.97e+02 at an offset of 118.338 usec -> 2018-04-02T19:43:10.948.769.087 (XO: 2.9489111)
   Min dTM/dt [1/us]::-3.68e+02 at an offset of 154.601 usec -> 2018-04-02T19:43:10.948.805.350 (XO: 2.9489474)

 - Block number: 2 starting at 2018-04-02T19:43:32.671.853.750 (XO: 24.6719959, t=35.6°C)  Rank: 00006
   TM values: Max =    31  Min =   -39  Ave =    -4.34  Std =     8.85
   Min TM value:-39 at an offset of 118.625 usec -> 2018-04-02T19:43:32.671.972.375 (XO: 24.6721145)
   Max TM value:31 at an offset of 118.688 usec -> 2018-04-02T19:43:32.671.972.437 (XO: 24.6721145)
   Max dTM/dt [1/us]::4.83e+02 at an offset of 118.663 usec -> 2018-04-02T19:43:32.671.972.412 (XO: 24.6721145)
   Min dTM/dt [1/us]::-4.61e+02 at an offset of 105.850 usec -> 2018-04-02T19:43:32.671.959.600 (XO: 24.6721017)

 - Block number: 3 starting at 2018-04-02T19:43:48.283.337.750 (XO: 40.2834797, t=35.6°C)  Rank: 00007
   TM values: Max =    22  Min =   -33  Ave =    -4.23  Std =     7.77
   Min TM value:-33 at an offset of 151.226 usec -> 2018-04-02T19:43:48.283.488.975 (XO: 40.2836309)
   Max TM value:22 at an offset of 150.326 usec -> 2018-04-02T19:43:48.283.488.075 (XO: 40.2836300)
   Max dTM/dt [1/us]::3.47e+02 at an offset of 153.276 usec -> 2018-04-02T19:43:48.283.491.025 (XO: 40.2836330)
   Min dTM/dt [1/us]::-3.92e+02 at an offset of 151.201 usec -> 2018-04-02T19:43:48.283.488.950 (XO: 40.2836309)
