-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180402_144043_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 3.482747 to 31.098837
-each separately from 3s to 31s and from 98548us to 483036us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.764639 in 1 packets - #/No: 18/18
      14.850584 in 1 packets - #/No: 36/18
MIU packet time from 2018-04-02T14:40:35.482.747.000 to 2018-04-02T14:41:03.098.837.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.764639 in 1 packets - #/No: 18/18
      14.850584 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.4342583 to 32.4341525
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.9999510 in 1 packets - #/No: 18/18
      14.9999432 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180402_144043_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.9170050 to 4.9172940
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.7645910  -- OK for SL --
   MIU time from 2018-04-02T14:40:35.482.747.000 to 2018-04-02T14:40:35.483.036.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.764639

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.6818850 to 17.6821740
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.8505270  -- OK for SL --
   MIU time from 2018-04-02T14:40:48.247.675.000 to 2018-04-02T14:40:48.247.964.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.850584

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.5327010 to 32.5329900
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-04-02T14:41:03.098.548.000 to 2018-04-02T14:41:03.098.837.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-04-02T14:40:35.482.606.750
   Max SL corrected packet time: 2018-04-02T14:41:03.098.696.750
   Min SL corrected XO rel time: 3.4827485
   Max SL corrected XO rel time: 31.0988379
   TM values from -16 to 7 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-04-02T14:40:35.482.606.750 (XO: 3.4827485, t=34.0°C)  Rank: 00002
   TM values: Max =     5  Min =   -13  Ave =    -4.48  Std =     2.69
   Min TM value:-13 at an offset of 106.150 usec -> 2018-04-02T14:40:35.482.712.900 (XO: 3.4828546)
   Max TM value:5 at an offset of 276.126 usec -> 2018-04-02T14:40:35.482.882.876 (XO: 3.4830246)
   Max dTM/dt [1/us]::7.40e+01 at an offset of 167.451 usec -> 2018-04-02T14:40:35.482.774.200 (XO: 3.4829159)
   Min dTM/dt [1/us]::-6.90e+01 at an offset of 297.189 usec -> 2018-04-02T14:40:35.482.903.938 (XO: 3.4830457)

 - Block number: 2 starting at 2018-04-02T14:40:48.247.534.750 (XO: 16.2476766, t=34.0°C)  Rank: 00002
   TM values: Max =     6  Min =   -16  Ave =    -4.33  Std =     3.49
   Min TM value:-16 at an offset of 129.100 usec -> 2018-04-02T14:40:48.247.663.850 (XO: 16.2478057)
   Max TM value:6 at an offset of 67.588 usec -> 2018-04-02T14:40:48.247.602.337 (XO: 16.2477442)
   Max dTM/dt [1/us]::9.20e+01 at an offset of 246.726 usec -> 2018-04-02T14:40:48.247.781.475 (XO: 16.2479234)
   Min dTM/dt [1/us]::-8.90e+01 at an offset of 87.613 usec -> 2018-04-02T14:40:48.247.622.362 (XO: 16.2477643)

 - Block number: 3 starting at 2018-04-02T14:41:03.098.407.750 (XO: 31.0985489, t=34.0°C)  Rank: 00002
   TM values: Max =     7  Min =   -15  Ave =    -4.33  Std =     3.21
   Min TM value:-15 at an offset of 124.788 usec -> 2018-04-02T14:41:03.098.532.537 (XO: 31.0986737)
   Max TM value:7 at an offset of 142.138 usec -> 2018-04-02T14:41:03.098.549.888 (XO: 31.0986910)
   Max dTM/dt [1/us]::8.40e+01 at an offset of 114.300 usec -> 2018-04-02T14:41:03.098.522.050 (XO: 31.0986632)
   Min dTM/dt [1/us]::-8.30e+01 at an offset of 92.063 usec -> 2018-04-02T14:41:03.098.499.812 (XO: 31.0986409)
