-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180326_224214_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 8.338202 to 30.050558
-each separately from 8s to 30s and from 50269us to 338491us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.734185 in 1 packets - #/No: 18/18
      14.977304 in 1 packets - #/No: 36/18
MIU packet time from 2018-03-26T22:42:10.338.202.000 to 2018-03-26T22:42:32.050.558.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.734185 in 1 packets - #/No: 18/18
      14.977304 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.4374354 to 31.4373516
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.9999733 in 1 packets - #/No: 18/18
      14.9999429 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      05 -> 48.5°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180326_224214_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.7756375 to 9.7759265
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.7341590  -- OK for SL --
   MIU time from 2018-03-26T22:42:10.338.202.000 to 2018-03-26T22:42:10.338.491.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.734185

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.5100855 to 16.5103745
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.9772470  -- OK for SL --
   MIU time from 2018-03-26T22:42:17.072.676.000 to 2018-03-26T22:42:17.072.965.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.977304

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.4876215 to 31.4879105
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-03-26T22:42:32.050.269.000 to 2018-03-26T22:42:32.050.558.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-03-26T22:42:10.338.061.750
   Max SL corrected packet time: 2018-03-26T22:42:32.050.417.750
   Min SL corrected XO rel time: 8.3382034
   Max SL corrected XO rel time: 30.0505591
   TM values from -30 to 26 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-03-26T22:42:10.338.061.750 (XO: 8.3382034, t=34.0°C)  Rank: 00007
   TM values: Max =    26  Min =   -30  Ave =    -4.47  Std =     6.53
   Min TM value:-30 at an offset of 145.338 usec -> 2018-03-26T22:42:10.338.207.088 (XO: 8.3383487)
   Max TM value:26 at an offset of 144.726 usec -> 2018-03-26T22:42:10.338.206.475 (XO: 8.3383481)
   Max dTM/dt [1/us]::2.44e+02 at an offset of 145.513 usec -> 2018-03-26T22:42:10.338.207.263 (XO: 8.3383489)
   Min dTM/dt [1/us]::-2.57e+02 at an offset of 123.138 usec -> 2018-03-26T22:42:10.338.184.887 (XO: 8.3383265)

 - Block number: 2 starting at 2018-03-26T22:42:17.072.535.750 (XO: 15.0726771, t=34.0°C)  Rank: 00002
   TM values: Max =     8  Min =   -16  Ave =    -4.40  Std =     3.18
   Min TM value:-16 at an offset of 178.588 usec -> 2018-03-26T22:42:17.072.714.338 (XO: 15.0728557)
   Max TM value:8 at an offset of 4.550 usec -> 2018-03-26T22:42:17.072.540.300 (XO: 15.0726816)
   Max dTM/dt [1/us]::1.16e+02 at an offset of 226.426 usec -> 2018-03-26T22:42:17.072.762.175 (XO: 15.0729035)
   Min dTM/dt [1/us]::-1.15e+02 at an offset of 99.625 usec -> 2018-03-26T22:42:17.072.635.375 (XO: 15.0727767)

 - Block number: 3 starting at 2018-03-26T22:42:32.050.128.750 (XO: 30.0502701, t=34.0°C)  Rank: 00002
   TM values: Max =     7  Min =   -14  Ave =    -4.45  Std =     2.76
   Min TM value:-14 at an offset of 96.288 usec -> 2018-03-26T22:42:32.050.225.037 (XO: 30.0503664)
   Max TM value:7 at an offset of 24.475 usec -> 2018-03-26T22:42:32.050.153.225 (XO: 30.0502946)
   Max dTM/dt [1/us]::1.14e+02 at an offset of 228.826 usec -> 2018-03-26T22:42:32.050.357.575 (XO: 30.0504989)
   Min dTM/dt [1/us]::-1.30e+02 at an offset of 284.926 usec -> 2018-03-26T22:42:32.050.413.676 (XO: 30.0505550)
