-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180324_062132_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 8.409729 to 40.361789
-each separately from 8s to 40s and from 361500us to 801451us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      9.391144 in 1 packets - #/No: 18/18
      22.560049 in 1 packets - #/No: 36/18
MIU packet time from 2018-03-24T06:21:27.505.729.000 to 2018-03-24T06:21:59.457.789.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      9.391144 in 1 packets - #/No: 18/18
      22.560049 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.4377221 to 41.4375832
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.9999609 in 1 packets - #/No: 18/18
      22.9999002 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      06 -> 47.2°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180324_062132_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      13 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.8474511 to 9.8477401
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.3911030  -- OK for SL --
   MIU time from 2018-03-24T06:21:27.505.729.000 to 2018-03-24T06:21:27.506.018.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.391144

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.2388431 to 19.2391321
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.5599510  -- OK for SL --
   MIU time from 2018-03-24T06:21:36.897.162.000 to 2018-03-24T06:21:36.897.451.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.560049

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      12 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.7990831 to 41.7993721
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-03-24T06:21:59.457.500.000 to 2018-03-24T06:21:59.457.789.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-03-24T06:21:27.505.588.750
   Max SL corrected packet time: 2018-03-24T06:21:59.457.648.750
   Min SL corrected XO rel time: 8.4097308
   Max SL corrected XO rel time: 40.3617905
   TM values from -34 to 26 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-03-24T06:21:27.505.588.750 (XO: 8.4097308, t=34.0°C)  Rank: 00013
   TM values: Max =    25  Min =   -32  Ave =    -4.35  Std =    10.36
   Min TM value:-32 at an offset of 199.188 usec -> 2018-03-24T06:21:27.505.787.938 (XO: 8.4099300)
   Max TM value:25 at an offset of 224.076 usec -> 2018-03-24T06:21:27.505.812.825 (XO: 8.4099549)
   Max dTM/dt [1/us]::4.95e+02 at an offset of 228.589 usec -> 2018-03-24T06:21:27.505.817.338 (XO: 8.4099594)
   Min dTM/dt [1/us]::-5.03e+02 at an offset of 229.476 usec -> 2018-03-24T06:21:27.505.818.226 (XO: 8.4099603)

 - Block number: 2 starting at 2018-03-24T06:21:36.897.021.750 (XO: 17.8011636, t=34.0°C)  Rank: 00008
   TM values: Max =    20  Min =   -30  Ave =    -4.39  Std =     6.81
   Min TM value:-30 at an offset of 212.701 usec -> 2018-03-24T06:21:36.897.234.450 (XO: 17.8013763)
   Max TM value:20 at an offset of 113.188 usec -> 2018-03-24T06:21:36.897.134.937 (XO: 17.8012768)
   Max dTM/dt [1/us]::3.86e+02 at an offset of 111.675 usec -> 2018-03-24T06:21:36.897.133.425 (XO: 17.8012753)
   Min dTM/dt [1/us]::-3.93e+02 at an offset of 111.925 usec -> 2018-03-24T06:21:36.897.133.675 (XO: 17.8012756)

 - Block number: 3 starting at 2018-03-24T06:21:59.457.359.750 (XO: 40.3615015, t=34.0°C)  Rank: 00012
   TM values: Max =    26  Min =   -34  Ave =    -4.34  Std =    11.27
   Min TM value:-34 at an offset of 115.150 usec -> 2018-03-24T06:21:59.457.474.900 (XO: 40.3616166)
   Max TM value:26 at an offset of 120.088 usec -> 2018-03-24T06:21:59.457.479.838 (XO: 40.3616215)
   Max dTM/dt [1/us]::5.47e+02 at an offset of 120.051 usec -> 2018-03-24T06:21:59.457.479.800 (XO: 40.3616215)
   Min dTM/dt [1/us]::-5.66e+02 at an offset of 120.951 usec -> 2018-03-24T06:21:59.457.480.700 (XO: 40.3616224)
