-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180313_222655_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 13.950048 to 32.050103
-each separately from 13s to 32s and from 49814us to 950337us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.971870 in 1 packets - #/No: 18/18
      10.127318 in 1 packets - #/No: 36/18
MIU packet time from 2018-03-13T22:26:55.950.048.000 to 2018-03-13T22:27:14.050.103.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.971870 in 1 packets - #/No: 18/18
      10.127318 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.4309727 to 33.4308998
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.9999692 in 1 packets - #/No: 18/18
      10.9999579 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180313_222655_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.3810191 to 15.3813081
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.9718390  -- OK for SL --
   MIU time from 2018-03-13T22:26:55.950.048.000 to 2018-03-13T22:26:55.950.337.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.971870

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      12 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.3531471 to 23.3534361
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.1272790  -- OK for SL --
   MIU time from 2018-03-13T22:27:03.922.207.000 to 2018-03-13T22:27:03.922.496.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.127318

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.4807151 to 33.4810041
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-03-13T22:27:14.049.814.000 to 2018-03-13T22:27:14.050.103.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-03-13T22:26:55.949.907.750
   Max SL corrected packet time: 2018-03-13T22:27:14.049.962.750
   Min SL corrected XO rel time: 13.9500501
   Max SL corrected XO rel time: 32.0501045
   TM values from -36 to 26 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-03-13T22:26:55.949.907.750 (XO: 13.9500501, t=34.0°C)  Rank: 00008
   TM values: Max =    22  Min =   -32  Ave =    -4.51  Std =     8.66
   Min TM value:-32 at an offset of 164.713 usec -> 2018-03-13T22:26:55.950.072.463 (XO: 13.9502148)
   Max TM value:22 at an offset of 118.013 usec -> 2018-03-13T22:26:55.950.025.762 (XO: 13.9501681)
   Max dTM/dt [1/us]::4.15e+02 at an offset of 154.401 usec -> 2018-03-13T22:26:55.950.062.150 (XO: 13.9502045)
   Min dTM/dt [1/us]::-4.25e+02 at an offset of 135.863 usec -> 2018-03-13T22:26:55.950.043.613 (XO: 13.9501860)

 - Block number: 2 starting at 2018-03-13T22:27:03.922.066.750 (XO: 21.9222087, t=34.0°C)  Rank: 00012
   TM values: Max =    26  Min =   -36  Ave =    -4.50  Std =    10.86
   Min TM value:-36 at an offset of 140.063 usec -> 2018-03-13T22:27:03.922.206.813 (XO: 21.9223488)
   Max TM value:26 at an offset of 143.876 usec -> 2018-03-13T22:27:03.922.210.625 (XO: 21.9223526)
   Max dTM/dt [1/us]::5.21e+02 at an offset of 146.588 usec -> 2018-03-13T22:27:03.922.213.338 (XO: 21.9223553)
   Min dTM/dt [1/us]::-5.07e+02 at an offset of 159.613 usec -> 2018-03-13T22:27:03.922.226.363 (XO: 21.9223683)

 - Block number: 3 starting at 2018-03-13T22:27:14.049.673.750 (XO: 32.0498155, t=34.0°C)  Rank: 00009
   TM values: Max =    23  Min =   -32  Ave =    -4.50  Std =     8.17
   Min TM value:-32 at an offset of 144.938 usec -> 2018-03-13T22:27:14.049.818.688 (XO: 32.0499604)
   Max TM value:23 at an offset of 144.051 usec -> 2018-03-13T22:27:14.049.817.800 (XO: 32.0499595)
   Max dTM/dt [1/us]::4.73e+02 at an offset of 144.013 usec -> 2018-03-13T22:27:14.049.817.763 (XO: 32.0499595)
   Min dTM/dt [1/us]::-4.42e+02 at an offset of 144.901 usec -> 2018-03-13T22:27:14.049.818.650 (XO: 32.0499604)
