-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180309_054134_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 13.781132 to 38.589832
-each separately from 13s to 38s and from 231265us to 781421us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      1.449844 in 1 packets - #/No: 18/18
      23.357989 in 1 packets - #/No: 36/18
MIU packet time from 2018-03-09T05:41:33.781.132.000 to 2018-03-09T05:41:58.589.832.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      1.449844 in 1 packets - #/No: 18/18
      23.357989 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.4369519 to 39.4368604
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      1.9999927 in 1 packets - #/No: 18/18
      22.9999158 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180309_054134_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      10 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.2180825 to 15.2183715
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 1.4498390  -- OK for SL --
   MIU time from 2018-03-09T05:41:33.781.132.000 to 2018-03-09T05:41:33.781.421.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 1.449844

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      11 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.6682105 to 16.6684995
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.3579030  -- OK for SL --
   MIU time from 2018-03-09T05:41:35.231.265.000 to 2018-03-09T05:41:35.231.554.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.357989

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      10 in 18 packets (all packets)  -- OK --
   10 MHz counter from 40.0264025 to 40.0266915
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-03-09T05:41:58.589.543.000 to 2018-03-09T05:41:58.589.832.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-03-09T05:41:33.780.991.750
   Max SL corrected packet time: 2018-03-09T05:41:58.589.691.750
   Min SL corrected XO rel time: 13.7811334
   Max SL corrected XO rel time: 38.5898332
   TM values from -35 to 25 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-03-09T05:41:33.780.991.750 (XO: 13.7811334, t=34.0°C)  Rank: 00010
   TM values: Max =    24  Min =   -33  Ave =    -4.45  Std =     9.45
   Min TM value:-33 at an offset of 166.551 usec -> 2018-03-09T05:41:33.781.158.300 (XO: 13.7813000)
   Max TM value:24 at an offset of 158.051 usec -> 2018-03-09T05:41:33.781.149.800 (XO: 13.7812915)
   Max dTM/dt [1/us]::5.25e+02 at an offset of 130.288 usec -> 2018-03-09T05:41:33.781.122.037 (XO: 13.7812637)
   Min dTM/dt [1/us]::-4.87e+02 at an offset of 129.700 usec -> 2018-03-09T05:41:33.781.121.450 (XO: 13.7812631)

 - Block number: 2 starting at 2018-03-09T05:41:35.231.124.750 (XO: 15.2312668, t=34.0°C)  Rank: 00011
   TM values: Max =    25  Min =   -33  Ave =    -4.48  Std =    10.50
   Min TM value:-33 at an offset of 127.575 usec -> 2018-03-09T05:41:35.231.252.325 (XO: 15.2313943)
   Max TM value:25 at an offset of 125.013 usec -> 2018-03-09T05:41:35.231.249.762 (XO: 15.2313918)
   Max dTM/dt [1/us]::4.92e+02 at an offset of 125.800 usec -> 2018-03-09T05:41:35.231.250.550 (XO: 15.2313926)
   Min dTM/dt [1/us]::-4.88e+02 at an offset of 125.713 usec -> 2018-03-09T05:41:35.231.250.462 (XO: 15.2313925)

 - Block number: 3 starting at 2018-03-09T05:41:58.589.402.750 (XO: 38.5895442, t=34.0°C)  Rank: 00010
   TM values: Max =    24  Min =   -35  Ave =    -4.58  Std =     8.62
   Min TM value:-35 at an offset of 157.301 usec -> 2018-03-09T05:41:58.589.560.050 (XO: 38.5897015)
   Max TM value:24 at an offset of 112.500 usec -> 2018-03-09T05:41:58.589.515.250 (XO: 38.5896567)
   Max dTM/dt [1/us]::4.72e+02 at an offset of 113.288 usec -> 2018-03-09T05:41:58.589.516.037 (XO: 38.5896575)
   Min dTM/dt [1/us]::-4.86e+02 at an offset of 157.263 usec -> 2018-03-09T05:41:58.589.560.013 (XO: 38.5897015)
