-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180304_202432_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 13.137747 to 39.911290
-each separately from 13s to 39s and from 137747us to 911290us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      5.681924 in 1 packets - #/No: 18/18
      21.090752 in 1 packets - #/No: 36/18
MIU packet time from 2018-03-04T20:24:30.137.747.000 to 2018-03-04T20:24:56.911.290.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      5.681924 in 1 packets - #/No: 18/18
      21.090752 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.4311126 to 40.4310129
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      4.9999808 in 1 packets - #/No: 18/18
      20.9999195 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180304_202432_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.5688602 to 14.5691492
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.6819030  -- OK for SL --
   MIU time from 2018-03-04T20:24:30.137.747.000 to 2018-03-04T20:24:30.138.036.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.681924

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.2510522 to 20.2513412
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.0906710  -- OK for SL --
   MIU time from 2018-03-04T20:24:35.819.960.000 to 2018-03-04T20:24:35.820.249.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.090752

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.3420122 to 41.3423012
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-03-04T20:24:56.911.001.000 to 2018-03-04T20:24:56.911.290.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-03-04T20:24:30.137.606.750
   Max SL corrected packet time: 2018-03-04T20:24:56.911.149.750
   Min SL corrected XO rel time: 13.1377481
   Max SL corrected XO rel time: 39.9112918
   TM values from -27 to 22 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-03-04T20:24:30.137.606.750 (XO: 13.1377481, t=34.0°C)  Rank: 00005
   TM values: Max =    22  Min =   -27  Ave =    -4.39  Std =     5.78
   Min TM value:-27 at an offset of 125.463 usec -> 2018-03-04T20:24:30.137.732.212 (XO: 13.1378736)
   Max TM value:22 at an offset of 121.550 usec -> 2018-03-04T20:24:30.137.728.300 (XO: 13.1378697)
   Max dTM/dt [1/us]::2.61e+02 at an offset of 232.351 usec -> 2018-03-04T20:24:30.137.839.100 (XO: 13.1379805)
   Min dTM/dt [1/us]::-2.87e+02 at an offset of 269.814 usec -> 2018-03-04T20:24:30.137.876.563 (XO: 13.1380180)

 - Block number: 2 starting at 2018-03-04T20:24:35.819.819.750 (XO: 18.8199619, t=34.0°C)  Rank: 00005
   TM values: Max =    20  Min =   -27  Ave =    -4.46  Std =     5.83
   Min TM value:-27 at an offset of 110.613 usec -> 2018-03-04T20:24:35.819.930.362 (XO: 18.8200725)
   Max TM value:20 at an offset of 113.388 usec -> 2018-03-04T20:24:35.819.933.137 (XO: 18.8200753)
   Max dTM/dt [1/us]::3.41e+02 at an offset of 111.175 usec -> 2018-03-04T20:24:35.819.930.925 (XO: 18.8200731)
   Min dTM/dt [1/us]::-3.19e+02 at an offset of 110.575 usec -> 2018-03-04T20:24:35.819.930.325 (XO: 18.8200725)

 - Block number: 3 starting at 2018-03-04T20:24:56.910.860.750 (XO: 39.9110028, t=34.0°C)  Rank: 00005
   TM values: Max =    17  Min =   -25  Ave =    -4.35  Std =     5.59
   Min TM value:-25 at an offset of 49.950 usec -> 2018-03-04T20:24:56.910.910.700 (XO: 39.9110527)
   Max TM value:17 at an offset of 53.875 usec -> 2018-03-04T20:24:56.910.914.625 (XO: 39.9110566)
   Max dTM/dt [1/us]::3.32e+02 at an offset of 148.138 usec -> 2018-03-04T20:24:56.911.008.888 (XO: 39.9111509)
   Min dTM/dt [1/us]::-3.21e+02 at an offset of 49.238 usec -> 2018-03-04T20:24:56.910.909.987 (XO: 39.9110520)
