-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180303_053740_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 3.497531 to 30.501234
-each separately from 3s to 30s and from 115591us to 501234us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      11.617771 in 1 packets - #/No: 18/18
      15.385065 in 1 packets - #/No: 36/18
MIU packet time from 2018-03-03T05:37:29.497.531.000 to 2018-03-03T05:37:56.501.234.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      11.617771 in 1 packets - #/No: 18/18
      15.385065 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.9390825 to 31.9389791
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      11.9999541 in 1 packets - #/No: 18/18
      14.9999425 in 1 packets - #/No: 36/18
Temperature FPGA: 
      07 -> 45.9°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      06 -> 47.2°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BE -> 32.4°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180303_053740_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.4366119 to 5.4369009
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.6177270  -- OK for SL --
   MIU time from 2018-03-03T05:37:29.497.531.000 to 2018-03-03T05:37:29.497.820.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.617771

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.0546279 to 17.0549169
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.3850070  -- OK for SL --
   MIU time from 2018-03-03T05:37:41.115.591.000 to 2018-03-03T05:37:41.115.880.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.385065

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.4399239 to 32.4402129
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-03-03T05:37:56.500.945.000 to 2018-03-03T05:37:56.501.234.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-03-03T05:37:29.497.390.750
   Max SL corrected packet time: 2018-03-03T05:37:56.501.093.750
   Min SL corrected XO rel time: 3.4975313
   Max SL corrected XO rel time: 30.5012357
   TM values from -20 to 11 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-03-03T05:37:29.497.390.750 (XO: 3.4975313, t=32.4°C)  Rank: 00002
   TM values: Max =    10  Min =   -17  Ave =    -4.44  Std =     3.78
   Min TM value:-17 at an offset of 20.850 usec -> 2018-03-03T05:37:29.497.411.600 (XO: 3.4975521)
   Max TM value:10 at an offset of 75.013 usec -> 2018-03-03T05:37:29.497.465.762 (XO: 3.4976063)
   Max dTM/dt [1/us]::1.51e+02 at an offset of 203.976 usec -> 2018-03-03T05:37:29.497.594.725 (XO: 3.4977353)
   Min dTM/dt [1/us]::-1.39e+02 at an offset of 14.013 usec -> 2018-03-03T05:37:29.497.404.762 (XO: 3.4975453)

 - Block number: 2 starting at 2018-03-03T05:37:41.115.450.750 (XO: 15.1155918, t=32.4°C)  Rank: 00003
   TM values: Max =    11  Min =   -20  Ave =    -4.44  Std =     4.82
   Min TM value:-20 at an offset of 100.800 usec -> 2018-03-03T05:37:41.115.551.550 (XO: 15.1156926)
   Max TM value:11 at an offset of 54.163 usec -> 2018-03-03T05:37:41.115.504.912 (XO: 15.1156459)
   Max dTM/dt [1/us]::1.82e+02 at an offset of 99.713 usec -> 2018-03-03T05:37:41.115.550.462 (XO: 15.1156915)
   Min dTM/dt [1/us]::-1.74e+02 at an offset of 107.800 usec -> 2018-03-03T05:37:41.115.558.550 (XO: 15.1156996)

 - Block number: 3 starting at 2018-03-03T05:37:56.500.804.750 (XO: 30.5009467, t=32.4°C)  Rank: 00002
   TM values: Max =     6  Min =   -16  Ave =    -4.42  Std =     2.97
   Min TM value:-16 at an offset of 79.075 usec -> 2018-03-03T05:37:56.500.883.825 (XO: 30.5010258)
   Max TM value:6 at an offset of 32.538 usec -> 2018-03-03T05:37:56.500.837.287 (XO: 30.5009792)
   Max dTM/dt [1/us]::9.80e+01 at an offset of 30.575 usec -> 2018-03-03T05:37:56.500.835.325 (XO: 30.5009773)
   Min dTM/dt [1/us]::-1.04e+02 at an offset of 192.776 usec -> 2018-03-03T05:37:56.500.997.525 (XO: 30.5011395)
