-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180302_132836_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.889189 to 33.410768
-each separately from 1s to 33s and from 315994us to 889478us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.426516 in 1 packets - #/No: 18/18
      18.094196 in 1 packets - #/No: 36/18
MIU packet time from 2018-03-02T13:28:22.889.189.000 to 2018-03-02T13:28:54.410.768.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.426516 in 1 packets - #/No: 18/18
      18.094196 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.4410965 to 34.4409732
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.9999459 in 1 packets - #/No: 18/18
      17.9999308 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      05 -> 48.5°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180302_132836_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.3302845 to 3.3305735
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.4264630  -- OK for SL --
   MIU time from 2018-03-02T13:28:22.889.189.000 to 2018-03-02T13:28:22.889.478.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.426516

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.7570365 to 16.7573255
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.0941270  -- OK for SL --
   MIU time from 2018-03-02T13:28:36.315.994.000 to 2018-03-02T13:28:36.316.283.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.094196

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.8514525 to 34.8517415
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-03-02T13:28:54.410.479.000 to 2018-03-02T13:28:54.410.768.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-03-02T13:28:22.889.048.750
   Max SL corrected packet time: 2018-03-02T13:28:54.410.627.750
   Min SL corrected XO rel time: 1.8891915
   Max SL corrected XO rel time: 33.4107699
   TM values from -21 to 13 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-03-02T13:28:22.889.048.750 (XO: 1.8891915, t=35.6°C)  Rank: 00004
   TM values: Max =    13  Min =   -21  Ave =    -4.31  Std =     6.61
   Min TM value:-21 at an offset of 111.788 usec -> 2018-03-02T13:28:22.889.160.537 (XO: 1.8893033)
   Max TM value:13 at an offset of 179.376 usec -> 2018-03-02T13:28:22.889.228.125 (XO: 1.8893708)
   Max dTM/dt [1/us]::1.28e+02 at an offset of 238.413 usec -> 2018-03-02T13:28:22.889.287.163 (XO: 1.8894299)
   Min dTM/dt [1/us]::-1.50e+02 at an offset of 305.989 usec -> 2018-03-02T13:28:22.889.354.738 (XO: 1.8894975)

 - Block number: 2 starting at 2018-03-02T13:28:36.315.853.750 (XO: 15.3159953, t=35.6°C)  Rank: 00002
   TM values: Max =     6  Min =   -15  Ave =    -4.37  Std =     3.22
   Min TM value:-15 at an offset of 213.313 usec -> 2018-03-02T13:28:36.316.067.063 (XO: 15.3162086)
   Max TM value:6 at an offset of 148.788 usec -> 2018-03-02T13:28:36.316.002.538 (XO: 15.3161441)
   Max dTM/dt [1/us]::6.70e+01 at an offset of 148.738 usec -> 2018-03-02T13:28:36.316.002.488 (XO: 15.3161440)
   Min dTM/dt [1/us]::-6.60e+01 at an offset of 140.176 usec -> 2018-03-02T13:28:36.315.993.925 (XO: 15.3161355)

 - Block number: 3 starting at 2018-03-02T13:28:54.410.338.750 (XO: 33.4104809, t=35.6°C)  Rank: 00003
   TM values: Max =     9  Min =   -17  Ave =    -4.29  Std =     3.86
   Min TM value:-17 at an offset of 88.500 usec -> 2018-03-02T13:28:54.410.427.250 (XO: 33.4105694)
   Max TM value:9 at an offset of 81.825 usec -> 2018-03-02T13:28:54.410.420.575 (XO: 33.4105627)
   Max dTM/dt [1/us]::8.60e+01 at an offset of 79.900 usec -> 2018-03-02T13:28:54.410.418.650 (XO: 33.4105608)
   Min dTM/dt [1/us]::-9.00e+01 at an offset of 13.738 usec -> 2018-03-02T13:28:54.410.352.487 (XO: 33.4104946)
