-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180228_223800_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.306321 to 35.467047
-each separately from 1s to 35s and from 78677us to 467047us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      21.772067 in 1 packets - #/No: 18/18
      12.387792 in 1 packets - #/No: 36/18
MIU packet time from 2018-02-28T22:37:46.306.321.000 to 2018-02-28T22:38:20.467.047.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      21.772067 in 1 packets - #/No: 18/18
      12.387792 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.4362794 to 36.4361469
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      21.9999144 in 1 packets - #/No: 18/18
      11.9999531 in 1 packets - #/No: 36/18
Temperature FPGA: 
      07 -> 45.9°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      06 -> 47.2°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      BE -> 32.4°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180228_223800_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.7426000 to 2.7428890
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.7719830  -- OK for SL --
   MIU time from 2018-02-28T22:37:46.306.321.000 to 2018-02-28T22:37:46.306.610.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.772067

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.5148720 to 24.5151610
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.3877430  -- OK for SL --
   MIU time from 2018-02-28T22:38:08.078.677.000 to 2018-02-28T22:38:08.078.966.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.387792

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.9029040 to 36.9031930
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-02-28T22:38:20.466.758.000 to 2018-02-28T22:38:20.467.047.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-02-28T22:37:46.306.180.750
   Max SL corrected packet time: 2018-02-28T22:38:20.466.906.750
   Min SL corrected XO rel time: 1.3063218
   Max SL corrected XO rel time: 35.4670479
   TM values from -31 to 21 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-02-28T22:37:46.306.180.750 (XO: 1.3063218, t=32.4°C)  Rank: 00004
   TM values: Max =    13  Min =   -25  Ave =    -4.50  Std =     5.24
   Min TM value:-25 at an offset of 272.564 usec -> 2018-02-28T22:37:46.306.453.313 (XO: 1.3065943)
   Max TM value:13 at an offset of 163.688 usec -> 2018-02-28T22:37:46.306.344.438 (XO: 1.3064855)
   Max dTM/dt [1/us]::1.72e+02 at an offset of 99.475 usec -> 2018-02-28T22:37:46.306.280.225 (XO: 1.3064212)
   Min dTM/dt [1/us]::-1.81e+02 at an offset of 164.138 usec -> 2018-02-28T22:37:46.306.344.888 (XO: 1.3064859)

 - Block number: 2 starting at 2018-02-28T22:38:08.078.536.750 (XO: 23.0786785, t=32.4°C)  Rank: 00007
   TM values: Max =    20  Min =   -26  Ave =    -4.35  Std =     6.66
   Min TM value:-26 at an offset of 84.863 usec -> 2018-02-28T22:38:08.078.621.612 (XO: 23.0787634)
   Max TM value:20 at an offset of 102.450 usec -> 2018-02-28T22:38:08.078.639.200 (XO: 23.0787810)
   Max dTM/dt [1/us]::2.61e+02 at an offset of 129.601 usec -> 2018-02-28T22:38:08.078.666.350 (XO: 23.0788081)
   Min dTM/dt [1/us]::-2.44e+02 at an offset of 84.838 usec -> 2018-02-28T22:38:08.078.621.587 (XO: 23.0787633)

 - Block number: 3 starting at 2018-02-28T22:38:20.466.617.750 (XO: 35.4667589, t=32.4°C)  Rank: 00006
   TM values: Max =    21  Min =   -31  Ave =    -4.42  Std =     7.57
   Min TM value:-31 at an offset of 115.588 usec -> 2018-02-28T22:38:20.466.733.337 (XO: 35.4668745)
   Max TM value:21 at an offset of 107.238 usec -> 2018-02-28T22:38:20.466.724.987 (XO: 35.4668662)
   Max dTM/dt [1/us]::2.67e+02 at an offset of 136.976 usec -> 2018-02-28T22:38:20.466.754.725 (XO: 35.4668959)
   Min dTM/dt [1/us]::-2.75e+02 at an offset of 124.375 usec -> 2018-02-28T22:38:20.466.742.125 (XO: 35.4668833)
