-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180228_113207_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 3.728552 to 34.792977
-each separately from 3s to 34s and from 99832us to 792977us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.370991 in 1 packets - #/No: 18/18
      10.692567 in 1 packets - #/No: 36/18
MIU packet time from 2018-02-28T11:31:54.827.552.000 to 2018-02-28T11:32:25.891.977.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.370991 in 1 packets - #/No: 18/18
      10.692567 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.4334142 to 35.4332943
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.9999187 in 1 packets - #/No: 18/18
      9.9999614 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      06 -> 47.2°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
      05 -> 48.5°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180228_113207_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.1619656 to 5.1622546
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.3709110  -- OK for SL --
   MIU time from 2018-02-28T11:31:54.827.552.000 to 2018-02-28T11:31:54.827.841.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.370991

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.5331656 to 25.5334546
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.6925270  -- OK for SL --
   MIU time from 2018-02-28T11:32:15.198.832.000 to 2018-02-28T11:32:15.199.121.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.692567

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.2259816 to 36.2262706
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-02-28T11:32:25.891.688.000 to 2018-02-28T11:32:25.891.977.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-02-28T11:31:54.827.411.750
   Max SL corrected packet time: 2018-02-28T11:32:25.891.836.750
   Min SL corrected XO rel time: 3.7285542
   Max SL corrected XO rel time: 34.7929794
   TM values from -21 to 17 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-02-28T11:31:54.827.411.750 (XO: 3.7285542, t=35.6°C)  Rank: 00002
   TM values: Max =     7  Min =   -17  Ave =    -4.33  Std =     3.08
   Min TM value:-17 at an offset of 125.375 usec -> 2018-02-28T11:31:54.827.537.125 (XO: 3.7286796)
   Max TM value:7 at an offset of 139.726 usec -> 2018-02-28T11:31:54.827.551.475 (XO: 3.7286940)
   Max dTM/dt [1/us]::1.02e+02 at an offset of 125.288 usec -> 2018-02-28T11:31:54.827.537.037 (XO: 3.7286795)
   Min dTM/dt [1/us]::-9.50e+01 at an offset of 135.326 usec -> 2018-02-28T11:31:54.827.547.075 (XO: 3.7286896)

 - Block number: 2 starting at 2018-02-28T11:32:15.198.691.750 (XO: 24.0998331, t=35.6°C)  Rank: 00002
   TM values: Max =    17  Min =   -21  Ave =    -4.45  Std =     3.64
   Min TM value:-21 at an offset of 97.875 usec -> 2018-02-28T11:32:15.198.789.625 (XO: 24.0999310)
   Max TM value:17 at an offset of 115.675 usec -> 2018-02-28T11:32:15.198.807.425 (XO: 24.0999488)
   Max dTM/dt [1/us]::1.23e+02 at an offset of 97.925 usec -> 2018-02-28T11:32:15.198.789.675 (XO: 24.0999310)
   Min dTM/dt [1/us]::-1.33e+02 at an offset of 113.900 usec -> 2018-02-28T11:32:15.198.805.650 (XO: 24.0999470)

 - Block number: 3 starting at 2018-02-28T11:32:25.891.547.750 (XO: 34.7926904, t=35.6°C)  Rank: 00002
   TM values: Max =     4  Min =   -14  Ave =    -4.41  Std =     2.72
   Min TM value:-14 at an offset of 61.363 usec -> 2018-02-28T11:32:25.891.609.112 (XO: 34.7927518)
   Max TM value:4 at an offset of 47.000 usec -> 2018-02-28T11:32:25.891.594.750 (XO: 34.7927374)
   Max dTM/dt [1/us]::7.70e+01 at an offset of 285.514 usec -> 2018-02-28T11:32:25.891.833.263 (XO: 34.7929759)
   Min dTM/dt [1/us]::-8.70e+01 at an offset of 155.513 usec -> 2018-02-28T11:32:25.891.703.263 (XO: 34.7928459)
