-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180227_111904_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.756234 to 33.470859
-each separately from 0s to 33s and from 470570us to 813665us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      22.056853 in 1 packets - #/No: 18/18
      10.656905 in 1 packets - #/No: 36/18
MIU packet time from 2018-02-27T11:18:49.756.234.000 to 2018-02-27T11:19:22.470.859.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      22.056853 in 1 packets - #/No: 18/18
      10.656905 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4374317 to 34.4373034
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      21.9999146 in 1 packets - #/No: 18/18
      10.9999571 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180227_111904_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.1936650 to 2.1939540
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.0567670  -- OK for SL --
   MIU time from 2018-02-27T11:18:49.756.234.000 to 2018-02-27T11:18:49.756.523.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.056853

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.2507210 to 24.2510100
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.6568630  -- OK for SL --
   MIU time from 2018-02-27T11:19:11.813.376.000 to 2018-02-27T11:19:11.813.665.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.656905

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.9078730 to 34.9081620
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-02-27T11:19:22.470.570.000 to 2018-02-27T11:19:22.470.859.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-02-27T11:18:49.756.093.750
   Max SL corrected packet time: 2018-02-27T11:19:22.470.718.750
   Min SL corrected XO rel time: 0.7562362
   Max SL corrected XO rel time: 33.4708604
   TM values from -22 to 12 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-02-27T11:18:49.756.093.750 (XO: 0.7562362, t=35.6°C)  Rank: 00002
   TM values: Max =    10  Min =   -18  Ave =    -4.52  Std =     4.12
   Min TM value:-18 at an offset of 113.850 usec -> 2018-02-27T11:18:49.756.207.600 (XO: 0.7563501)
   Max TM value:10 at an offset of 138.976 usec -> 2018-02-27T11:18:49.756.232.725 (XO: 0.7563752)
   Max dTM/dt [1/us]::1.10e+02 at an offset of 148.376 usec -> 2018-02-27T11:18:49.756.242.125 (XO: 0.7563846)
   Min dTM/dt [1/us]::-1.12e+02 at an offset of 163.838 usec -> 2018-02-27T11:18:49.756.257.588 (XO: 0.7564001)

 - Block number: 2 starting at 2018-02-27T11:19:11.813.235.750 (XO: 22.8133779, t=35.6°C)  Rank: 00003
   TM values: Max =    12  Min =   -21  Ave =    -4.44  Std =     5.00
   Min TM value:-21 at an offset of 163.188 usec -> 2018-02-27T11:19:11.813.398.938 (XO: 22.8135411)
   Max TM value:12 at an offset of 124.400 usec -> 2018-02-27T11:19:11.813.360.150 (XO: 22.8135023)
   Max dTM/dt [1/us]::1.31e+02 at an offset of 101.350 usec -> 2018-02-27T11:19:11.813.337.100 (XO: 22.8134792)
   Min dTM/dt [1/us]::-1.30e+02 at an offset of 99.175 usec -> 2018-02-27T11:19:11.813.334.925 (XO: 22.8134770)

 - Block number: 3 starting at 2018-02-27T11:19:22.470.429.750 (XO: 33.4705714, t=35.6°C)  Rank: 00004
   TM values: Max =    12  Min =   -22  Ave =    -4.49  Std =     5.26
   Min TM value:-22 at an offset of 134.438 usec -> 2018-02-27T11:19:22.470.564.188 (XO: 33.4707059)
   Max TM value:12 at an offset of 124.688 usec -> 2018-02-27T11:19:22.470.554.437 (XO: 33.4706961)
   Max dTM/dt [1/us]::1.31e+02 at an offset of 299.426 usec -> 2018-02-27T11:19:22.470.729.176 (XO: 33.4708709)
   Min dTM/dt [1/us]::-1.35e+02 at an offset of 134.001 usec -> 2018-02-27T11:19:22.470.563.750 (XO: 33.4707054)
