-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180227_094816_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 2.954162 to 31.240240
-each separately from 2s to 31s and from 239951us to 954451us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.586826 in 1 packets - #/No: 18/18
      15.698385 in 1 packets - #/No: 36/18
MIU packet time from 2018-02-27T09:48:03.954.162.000 to 2018-02-27T09:48:32.240.240.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.586826 in 1 packets - #/No: 18/18
      15.698385 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.4384364 to 32.4383731
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.9999716 in 1 packets - #/No: 18/18
      15.9999651 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180227_094816_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.3925964 to 4.3928854
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.5867990  -- OK for SL --
   MIU time from 2018-02-27T09:48:03.954.162.000 to 2018-02-27T09:48:03.954.451.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.586826

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.9796844 to 16.9799734
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.6983510  -- OK for SL --
   MIU time from 2018-02-27T09:48:16.541.277.000 to 2018-02-27T09:48:16.541.566.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.698385

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.6783244 to 32.6786134
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-02-27T09:48:32.239.951.000 to 2018-02-27T09:48:32.240.240.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-02-27T09:48:03.954.021.750
   Max SL corrected packet time: 2018-02-27T09:48:32.240.099.750
   Min SL corrected XO rel time: 2.9541621
   Max SL corrected XO rel time: 31.2402408
   TM values from -20 to 11 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-02-27T09:48:03.954.021.750 (XO: 2.9541621, t=35.6°C)  Rank: 00003
   TM values: Max =     9  Min =   -18  Ave =    -4.39  Std =     4.98
   Min TM value:-18 at an offset of 107.300 usec -> 2018-02-27T09:48:03.954.129.050 (XO: 2.9542694)
   Max TM value:9 at an offset of 95.600 usec -> 2018-02-27T09:48:03.954.117.350 (XO: 2.9542577)
   Max dTM/dt [1/us]::1.14e+02 at an offset of 167.150 usec -> 2018-02-27T09:48:03.954.188.900 (XO: 2.9543292)
   Min dTM/dt [1/us]::-1.06e+02 at an offset of 150.900 usec -> 2018-02-27T09:48:03.954.172.650 (XO: 2.9543130)

 - Block number: 2 starting at 2018-02-27T09:48:16.541.136.750 (XO: 15.5412776, t=35.6°C)  Rank: 00003
   TM values: Max =    11  Min =   -20  Ave =    -4.39  Std =     6.49
   Min TM value:-20 at an offset of 139.650 usec -> 2018-02-27T09:48:16.541.276.400 (XO: 15.5414172)
   Max TM value:11 at an offset of 129.713 usec -> 2018-02-27T09:48:16.541.266.462 (XO: 15.5414073)
   Max dTM/dt [1/us]::1.25e+02 at an offset of 228.101 usec -> 2018-02-27T09:48:16.541.364.850 (XO: 15.5415057)
   Min dTM/dt [1/us]::-1.30e+02 at an offset of 228.851 usec -> 2018-02-27T09:48:16.541.365.600 (XO: 15.5415064)

 - Block number: 3 starting at 2018-02-27T09:48:32.239.810.750 (XO: 31.2399518, t=35.6°C)  Rank: 00002
   TM values: Max =     7  Min =   -15  Ave =    -4.40  Std =     3.41
   Min TM value:-15 at an offset of 98.088 usec -> 2018-02-27T09:48:32.239.908.837 (XO: 31.2400499)
   Max TM value:7 at an offset of 215.975 usec -> 2018-02-27T09:48:32.240.026.725 (XO: 31.2401678)
   Max dTM/dt [1/us]::7.70e+01 at an offset of 244.651 usec -> 2018-02-27T09:48:32.240.055.400 (XO: 31.2401965)
   Min dTM/dt [1/us]::-7.90e+01 at an offset of 202.800 usec -> 2018-02-27T09:48:32.240.013.550 (XO: 31.2401546)
