-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180224_111057_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.423588 to 31.178906
-each separately from 1s to 31s and from 75946us to 423877us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.652069 in 1 packets - #/No: 18/18
      16.102382 in 1 packets - #/No: 36/18
MIU packet time from 2018-02-24T11:10:43.423.588.000 to 2018-02-24T11:11:13.178.906.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.652069 in 1 packets - #/No: 18/18
      16.102382 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.9408373 to 32.9407193
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.9999450 in 1 packets - #/No: 18/18
      15.9999370 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180224_111057_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.3644251 to 3.3647141
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.6520150  -- OK for SL --
   MIU time from 2018-02-24T11:10:43.423.588.000 to 2018-02-24T11:10:43.423.877.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.652069

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.0167291 to 17.0170181
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.1023190  -- OK for SL --
   MIU time from 2018-02-24T11:10:57.075.946.000 to 2018-02-24T11:10:57.076.235.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.102382

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.1193371 to 33.1196261
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-02-24T11:11:13.178.617.000 to 2018-02-24T11:11:13.178.906.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-02-24T11:10:43.423.447.750
   Max SL corrected packet time: 2018-02-24T11:11:13.178.765.750
   Min SL corrected XO rel time: 1.4235895
   Max SL corrected XO rel time: 31.1789075
   TM values from -18 to 9 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-02-24T11:10:43.423.447.750 (XO: 1.4235895, t=34.0°C)  Rank: 00002
   TM values: Max =     8  Min =   -17  Ave =    -4.50  Std =     4.30
   Min TM value:-17 at an offset of 171.463 usec -> 2018-02-24T11:10:43.423.619.213 (XO: 1.4237610)
   Max TM value:8 at an offset of 242.301 usec -> 2018-02-24T11:10:43.423.690.050 (XO: 1.4238318)
   Max dTM/dt [1/us]::1.04e+02 at an offset of 194.501 usec -> 2018-02-24T11:10:43.423.642.250 (XO: 1.4237840)
   Min dTM/dt [1/us]::-1.03e+02 at an offset of 219.688 usec -> 2018-02-24T11:10:43.423.667.438 (XO: 1.4238092)

 - Block number: 2 starting at 2018-02-24T11:10:57.075.805.750 (XO: 15.0759471, t=34.0°C)  Rank: 00002
   TM values: Max =     7  Min =   -17  Ave =    -4.48  Std =     4.28
   Min TM value:-17 at an offset of 122.663 usec -> 2018-02-24T11:10:57.075.928.412 (XO: 15.0760698)
   Max TM value:7 at an offset of 80.800 usec -> 2018-02-24T11:10:57.075.886.550 (XO: 15.0760279)
   Max dTM/dt [1/us]::1.15e+02 at an offset of 80.875 usec -> 2018-02-24T11:10:57.075.886.625 (XO: 15.0760280)
   Min dTM/dt [1/us]::-1.14e+02 at an offset of 305.989 usec -> 2018-02-24T11:10:57.076.111.738 (XO: 15.0762531)

 - Block number: 3 starting at 2018-02-24T11:11:13.178.476.750 (XO: 31.1786185, t=34.0°C)  Rank: 00002
   TM values: Max =     9  Min =   -18  Ave =    -4.54  Std =     4.04
   Min TM value:-18 at an offset of 84.863 usec -> 2018-02-24T11:11:13.178.561.612 (XO: 31.1787034)
   Max TM value:9 at an offset of 149.326 usec -> 2018-02-24T11:11:13.178.626.075 (XO: 31.1787678)
   Max dTM/dt [1/us]::1.07e+02 at an offset of 110.913 usec -> 2018-02-24T11:11:13.178.587.662 (XO: 31.1787294)
   Min dTM/dt [1/us]::-1.00e+02 at an offset of 110.025 usec -> 2018-02-24T11:11:13.178.586.775 (XO: 31.1787285)
