-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180223_125918_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.696866 to 41.839268
-each separately from 0s to 41s and from 15646us to 839268us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      23.318491 in 1 packets - #/No: 18/18
      17.823044 in 1 packets - #/No: 36/18
MIU packet time from 2018-02-23T12:59:02.696.866.000 to 2018-02-23T12:59:43.839.268.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      23.318491 in 1 packets - #/No: 18/18
      17.823044 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4368057 to 42.4366452
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      23.9999060 in 1 packets - #/No: 18/18
      16.9999335 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      06 -> 47.2°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180223_125918_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.1336703 to 2.1339593
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.3183990  -- OK for SL --
   MIU time from 2018-02-23T12:59:02.696.866.000 to 2018-02-23T12:59:02.697.155.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.318491

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.4523583 to 25.4526473
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.8229750  -- OK for SL --
   MIU time from 2018-02-23T12:59:26.015.646.000 to 2018-02-23T12:59:26.015.935.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.823044

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.2756223 to 43.2759113
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-02-23T12:59:43.838.979.000 to 2018-02-23T12:59:43.839.268.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-02-23T12:59:02.696.725.750
   Max SL corrected packet time: 2018-02-23T12:59:43.839.127.750
   Min SL corrected XO rel time: 0.6968674
   Max SL corrected XO rel time: 41.8392695
   TM values from -19 to 9 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-02-23T12:59:02.696.725.750 (XO: 0.6968674, t=34.0°C)  Rank: 00002
   TM values: Max =     9  Min =   -19  Ave =    -4.49  Std =     4.13
   Min TM value:-19 at an offset of 56.300 usec -> 2018-02-23T12:59:02.696.782.050 (XO: 0.6969237)
   Max TM value:9 at an offset of 286.939 usec -> 2018-02-23T12:59:02.697.012.688 (XO: 0.6971543)
   Max dTM/dt [1/us]::1.01e+02 at an offset of 256.226 usec -> 2018-02-23T12:59:02.696.981.976 (XO: 0.6971236)
   Min dTM/dt [1/us]::-1.02e+02 at an offset of 171.651 usec -> 2018-02-23T12:59:02.696.897.400 (XO: 0.6970390)

 - Block number: 2 starting at 2018-02-23T12:59:26.015.505.750 (XO: 24.0156467, t=34.0°C)  Rank: 00002
   TM values: Max =     6  Min =   -15  Ave =    -4.58  Std =     2.84
   Min TM value:-15 at an offset of 184.138 usec -> 2018-02-23T12:59:26.015.689.888 (XO: 24.0158308)
   Max TM value:6 at an offset of 110.113 usec -> 2018-02-23T12:59:26.015.615.862 (XO: 24.0157568)
   Max dTM/dt [1/us]::8.00e+01 at an offset of 20.600 usec -> 2018-02-23T12:59:26.015.526.350 (XO: 24.0156673)
   Min dTM/dt [1/us]::-7.40e+01 at an offset of 22.838 usec -> 2018-02-23T12:59:26.015.528.587 (XO: 24.0156695)

 - Block number: 3 starting at 2018-02-23T12:59:43.838.838.750 (XO: 41.8389805, t=34.0°C)  Rank: 00004
   TM values: Max =     5  Min =   -14  Ave =    -4.58  Std =     2.61
   Min TM value:-14 at an offset of 40.788 usec -> 2018-02-23T12:59:43.838.879.537 (XO: 41.8390212)
   Max TM value:5 at an offset of 24.725 usec -> 2018-02-23T12:59:43.838.863.475 (XO: 41.8390052)
   Max dTM/dt [1/us]::7.40e+01 at an offset of 159.338 usec -> 2018-02-23T12:59:43.838.998.088 (XO: 41.8391398)
   Min dTM/dt [1/us]::-8.90e+01 at an offset of 158.801 usec -> 2018-02-23T12:59:43.838.997.550 (XO: 41.8391393)
