-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180220_202534_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.622317 to 30.070385
-each separately from 0s to 30s and from 70096us to 812132us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      22.189237 in 1 packets - #/No: 18/18
      7.257964 in 1 packets - #/No: 36/18
MIU packet time from 2018-02-20T20:25:18.622.317.000 to 2018-02-20T20:25:48.070.385.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      22.189237 in 1 packets - #/No: 18/18
      7.257964 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4367183 to 31.4366013
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      21.9999143 in 1 packets - #/No: 18/18
      7.9999687 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      04 -> 49.8°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
      06 -> 47.2°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180220_202534_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.0590346 to 2.0593236
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.1891510  -- OK for SL --
   MIU time from 2018-02-20T20:25:18.622.317.000 to 2018-02-20T20:25:18.622.606.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.189237

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.2484746 to 24.2487636
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.2579350  -- OK for SL --
   MIU time from 2018-02-20T20:25:40.811.843.000 to 2018-02-20T20:25:40.812.132.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.257964

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.5066986 to 31.5069876
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-02-20T20:25:48.070.096.000 to 2018-02-20T20:25:48.070.385.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-02-20T20:25:18.622.176.750
   Max SL corrected packet time: 2018-02-20T20:25:48.070.244.750
   Min SL corrected XO rel time: 0.6223187
   Max SL corrected XO rel time: 30.0703866
   TM values from -22 to 15 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-02-20T20:25:18.622.176.750 (XO: 0.6223187, t=35.6°C)  Rank: 00003
   TM values: Max =    10  Min =   -19  Ave =    -4.47  Std =     3.84
   Min TM value:-19 at an offset of 115.738 usec -> 2018-02-20T20:25:18.622.292.487 (XO: 0.6224344)
   Max TM value:10 at an offset of 118.138 usec -> 2018-02-20T20:25:18.622.294.887 (XO: 0.6224368)
   Max dTM/dt [1/us]::1.68e+02 at an offset of 299.576 usec -> 2018-02-20T20:25:18.622.476.326 (XO: 0.6226182)
   Min dTM/dt [1/us]::-1.68e+02 at an offset of 174.763 usec -> 2018-02-20T20:25:18.622.351.513 (XO: 0.6224934)

 - Block number: 2 starting at 2018-02-20T20:25:40.811.702.750 (XO: 22.8118452, t=35.6°C)  Rank: 00004
   TM values: Max =    15  Min =   -22  Ave =    -4.41  Std =     4.39
   Min TM value:-22 at an offset of 117.513 usec -> 2018-02-20T20:25:40.811.820.262 (XO: 22.8119627)
   Max TM value:15 at an offset of 118.275 usec -> 2018-02-20T20:25:40.811.821.025 (XO: 22.8119634)
   Max dTM/dt [1/us]::2.47e+02 at an offset of 116.063 usec -> 2018-02-20T20:25:40.811.818.812 (XO: 22.8119612)
   Min dTM/dt [1/us]::-2.50e+02 at an offset of 122.988 usec -> 2018-02-20T20:25:40.811.825.737 (XO: 22.8119682)

 - Block number: 3 starting at 2018-02-20T20:25:48.069.955.750 (XO: 30.0700976, t=35.6°C)  Rank: 00003
   TM values: Max =    13  Min =   -20  Ave =    -4.48  Std =     4.57
   Min TM value:-20 at an offset of 133.551 usec -> 2018-02-20T20:25:48.070.089.300 (XO: 30.0702311)
   Max TM value:13 at an offset of 223.438 usec -> 2018-02-20T20:25:48.070.179.188 (XO: 30.0703210)
   Max dTM/dt [1/us]::1.98e+02 at an offset of 133.588 usec -> 2018-02-20T20:25:48.070.089.338 (XO: 30.0702312)
   Min dTM/dt [1/us]::-2.51e+02 at an offset of 135.001 usec -> 2018-02-20T20:25:48.070.090.750 (XO: 30.0702326)
