-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180218_115341_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.208462 to 31.044992
-each separately from 0s to 31s and from 44703us to 411579us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      18.202539 in 1 packets - #/No: 18/18
      12.633124 in 1 packets - #/No: 36/18
MIU packet time from 2018-02-18T11:53:25.208.462.000 to 2018-02-18T11:53:56.044.992.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      18.202539 in 1 packets - #/No: 18/18
      12.633124 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.9422996 to 32.9421699
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.9999247 in 1 packets - #/No: 18/18
      12.9999456 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      04 -> 49.8°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180218_115341_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.1507618 to 2.1510508
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.2024630  -- OK for SL --
   MIU time from 2018-02-18T11:53:25.208.462.000 to 2018-02-18T11:53:25.208.751.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.202539

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.3535138 to 20.3538028
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.6330710  -- OK for SL --
   MIU time from 2018-02-18T11:53:43.411.290.000 to 2018-02-18T11:53:43.411.579.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.633124

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.9868738 to 32.9871628
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-02-18T11:53:56.044.703.000 to 2018-02-18T11:53:56.044.992.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-02-18T11:53:25.208.321.750
   Max SL corrected packet time: 2018-02-18T11:53:56.044.851.750
   Min SL corrected XO rel time: 0.2084631
   Max SL corrected XO rel time: 31.0449931
   TM values from -20 to 13 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-02-18T11:53:25.208.321.750 (XO: 0.2084631, t=35.6°C)  Rank: 00002
   TM values: Max =     9  Min =   -17  Ave =    -4.36  Std =     3.31
   Min TM value:-17 at an offset of 177.388 usec -> 2018-02-18T11:53:25.208.499.138 (XO: 0.2086405)
   Max TM value:9 at an offset of 67.063 usec -> 2018-02-18T11:53:25.208.388.812 (XO: 0.2085301)
   Max dTM/dt [1/us]::1.17e+02 at an offset of 177.563 usec -> 2018-02-18T11:53:25.208.499.313 (XO: 0.2086406)
   Min dTM/dt [1/us]::-1.30e+02 at an offset of 178.313 usec -> 2018-02-18T11:53:25.208.500.063 (XO: 0.2086414)

 - Block number: 2 starting at 2018-02-18T11:53:43.411.149.750 (XO: 18.4112912, t=35.6°C)  Rank: 00003
   TM values: Max =    13  Min =   -20  Ave =    -4.42  Std =     5.26
   Min TM value:-20 at an offset of 169.701 usec -> 2018-02-18T11:53:43.411.319.450 (XO: 18.4114609)
   Max TM value:13 at an offset of 73.188 usec -> 2018-02-18T11:53:43.411.222.937 (XO: 18.4113644)
   Max dTM/dt [1/us]::1.53e+02 at an offset of 106.163 usec -> 2018-02-18T11:53:43.411.255.912 (XO: 18.4113974)
   Min dTM/dt [1/us]::-1.56e+02 at an offset of 170.501 usec -> 2018-02-18T11:53:43.411.320.250 (XO: 18.4114617)

 - Block number: 3 starting at 2018-02-18T11:53:56.044.562.750 (XO: 31.0447041, t=35.6°C)  Rank: 00002
   TM values: Max =     8  Min =   -17  Ave =    -4.34  Std =     3.37
   Min TM value:-17 at an offset of 105.500 usec -> 2018-02-18T11:53:56.044.668.250 (XO: 31.0448096)
   Max TM value:8 at an offset of 92.500 usec -> 2018-02-18T11:53:56.044.655.250 (XO: 31.0447966)
   Max dTM/dt [1/us]::9.70e+01 at an offset of 143.576 usec -> 2018-02-18T11:53:56.044.706.325 (XO: 31.0448477)
   Min dTM/dt [1/us]::-9.20e+01 at an offset of 173.701 usec -> 2018-02-18T11:53:56.044.736.450 (XO: 31.0448778)
