-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180211_141227_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 12.817819 to 41.662334
-each separately from 12s to 41s and from 467526us to 818108us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.649418 in 1 packets - #/No: 18/18
      22.194230 in 1 packets - #/No: 36/18
MIU packet time from 2018-02-11T14:12:23.817.819.000 to 2018-02-11T14:12:52.662.334.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.649418 in 1 packets - #/No: 18/18
      22.194230 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.9391935 to 42.9390790
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.9999723 in 1 packets - #/No: 18/18
      21.9999132 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      C0 -> 35.6°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180211_141227_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.7570111 to 14.7573001
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.6493910  -- OK for SL --
   MIU time from 2018-02-11T14:12:23.817.819.000 to 2018-02-11T14:12:23.818.108.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.649418

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.4066911 to 21.4069801
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.1941430  -- OK for SL --
   MIU time from 2018-02-11T14:12:30.467.526.000 to 2018-02-11T14:12:30.467.815.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.194230

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.6011231 to 43.6014121
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-02-11T14:12:52.662.045.000 to 2018-02-11T14:12:52.662.334.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-02-11T14:12:23.817.678.750
   Max SL corrected packet time: 2018-02-11T14:12:52.662.193.750
   Min SL corrected XO rel time: 12.8178209
   Max SL corrected XO rel time: 41.6623357
   TM values from -17 to 10 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-02-11T14:12:23.817.678.750 (XO: 12.8178209, t=34.0°C)  Rank: 00002
   TM values: Max =     6  Min =   -15  Ave =    -4.40  Std =     3.07
   Min TM value:-15 at an offset of 58.813 usec -> 2018-02-11T14:12:23.817.737.562 (XO: 12.8178797)
   Max TM value:6 at an offset of 50.413 usec -> 2018-02-11T14:12:23.817.729.162 (XO: 12.8178713)
   Max dTM/dt [1/us]::7.60e+01 at an offset of 28.625 usec -> 2018-02-11T14:12:23.817.707.375 (XO: 12.8178495)
   Min dTM/dt [1/us]::-7.60e+01 at an offset of 152.638 usec -> 2018-02-11T14:12:23.817.831.388 (XO: 12.8179735)

 - Block number: 2 starting at 2018-02-11T14:12:30.467.385.750 (XO: 19.4675271, t=34.0°C)  Rank: 00002
   TM values: Max =     6  Min =   -17  Ave =    -4.33  Std =     3.03
   Min TM value:-17 at an offset of 134.888 usec -> 2018-02-11T14:12:30.467.520.638 (XO: 19.4676620)
   Max TM value:6 at an offset of 32.225 usec -> 2018-02-11T14:12:30.467.417.975 (XO: 19.4675593)
   Max dTM/dt [1/us]::8.00e+01 at an offset of 68.163 usec -> 2018-02-11T14:12:30.467.453.912 (XO: 19.4675953)
   Min dTM/dt [1/us]::-9.30e+01 at an offset of 134.776 usec -> 2018-02-11T14:12:30.467.520.525 (XO: 19.4676619)

 - Block number: 3 starting at 2018-02-11T14:12:52.661.904.750 (XO: 41.6620467, t=34.0°C)  Rank: 00005
   TM values: Max =    10  Min =   -17  Ave =    -4.30  Std =     3.35
   Min TM value:-17 at an offset of 87.075 usec -> 2018-02-11T14:12:52.661.991.825 (XO: 41.6621338)
   Max TM value:10 at an offset of 92.438 usec -> 2018-02-11T14:12:52.661.997.187 (XO: 41.6621392)
   Max dTM/dt [1/us]::8.20e+01 at an offset of 231.838 usec -> 2018-02-11T14:12:52.662.136.588 (XO: 41.6622786)
   Min dTM/dt [1/us]::-9.60e+01 at an offset of 223.613 usec -> 2018-02-11T14:12:52.662.128.363 (XO: 41.6622704)
