-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180103_163057_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.003398 to 30.197542
-each separately from 0s to 30s and from 3398us to 197542us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.024686 in 1 packets - #/No: 18/18
      15.168591 in 1 packets - #/No: 36/18
MIU packet time from 2018-01-03T16:31:00.003.398.000 to 2018-01-03T16:31:30.197.542.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.024686 in 1 packets - #/No: 18/18
      15.168591 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4398713 to 31.4397444
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999365 in 1 packets - #/No: 18/18
      14.9999366 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      BE -> 32.4°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180103_163057_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4432702 to 1.4435592
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0246230  -- OK for SL --
   MIU time from 2018-01-03T16:31:00.003.398.000 to 2018-01-03T16:31:00.003.687.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.024686

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4681822 to 16.4684712
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.1685270  -- OK for SL --
   MIU time from 2018-01-03T16:31:15.028.373.000 to 2018-01-03T16:31:15.028.662.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.168591

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.6369982 to 31.6372872
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-01-03T16:31:30.197.253.000 to 2018-01-03T16:31:30.197.542.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-01-03T16:31:00.003.257.750
   Max SL corrected packet time: 2018-01-03T16:31:30.197.401.750
   Min SL corrected XO rel time: 0.0033989
   Max SL corrected XO rel time: 30.1975436
   TM values from -14 to 6 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-01-03T16:31:00.003.257.750 (XO: 0.0033989, t=34.0°C)  Rank: 00002
   TM values: Max =     6  Min =   -14  Ave =    -4.34  Std =     3.01
   Min TM value:-14 at an offset of 155.126 usec -> 2018-01-03T16:31:00.003.412.875 (XO: 0.0035540)
   Max TM value:6 at an offset of 89.050 usec -> 2018-01-03T16:31:00.003.346.800 (XO: 0.0034880)
   Max dTM/dt [1/us]::1.16e+02 at an offset of 94.725 usec -> 2018-01-03T16:31:00.003.352.475 (XO: 0.0034936)
   Min dTM/dt [1/us]::-1.26e+02 at an offset of 34.500 usec -> 2018-01-03T16:31:00.003.292.250 (XO: 0.0034334)

 - Block number: 2 starting at 2018-01-03T16:31:15.028.232.750 (XO: 15.0283745, t=34.0°C)  Rank: 00002
   TM values: Max =     6  Min =   -14  Ave =    -4.32  Std =     2.77
   Min TM value:-14 at an offset of 117.200 usec -> 2018-01-03T16:31:15.028.349.950 (XO: 15.0284917)
   Max TM value:6 at an offset of 64.638 usec -> 2018-01-03T16:31:15.028.297.387 (XO: 15.0284392)
   Max dTM/dt [1/us]::9.10e+01 at an offset of 54.925 usec -> 2018-01-03T16:31:15.028.287.675 (XO: 15.0284294)
   Min dTM/dt [1/us]::-1.00e+02 at an offset of 287.114 usec -> 2018-01-03T16:31:15.028.519.863 (XO: 15.0286616)

 - Block number: 3 starting at 2018-01-03T16:31:30.197.112.750 (XO: 30.1972546, t=34.0°C)  Rank: 00002
   TM values: Max =     5  Min =   -14  Ave =    -4.35  Std =     3.03
   Min TM value:-14 at an offset of 139.838 usec -> 2018-01-03T16:31:30.197.252.588 (XO: 30.1973945)
   Max TM value:5 at an offset of 15.713 usec -> 2018-01-03T16:31:30.197.128.462 (XO: 30.1972703)
   Max dTM/dt [1/us]::8.70e+01 at an offset of 42.700 usec -> 2018-01-03T16:31:30.197.155.450 (XO: 30.1972973)
   Min dTM/dt [1/us]::-9.20e+01 at an offset of 160.726 usec -> 2018-01-03T16:31:30.197.273.475 (XO: 30.1974154)
