-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180103_005241_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.614287 to 32.039405
-each separately from 1s to 32s and from 39116us to 801491us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.186626 in 1 packets - #/No: 18/18
      10.237625 in 1 packets - #/No: 36/18
MIU packet time from 2018-01-03T00:52:45.614.287.000 to 2018-01-03T00:53:16.039.405.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.186626 in 1 packets - #/No: 18/18
      10.237625 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.4376341 to 33.4375072
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      19.9999181 in 1 packets - #/No: 18/18
      10.9999550 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      07 -> 45.9°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BE -> 32.4°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180103_005241_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.0519202 to 3.0522092
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.1865430  -- OK for SL --
   MIU time from 2018-01-03T00:52:45.614.287.000 to 2018-01-03T00:52:45.614.576.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.186626

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.2387522 to 23.2390412
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.2375830  -- OK for SL --
   MIU time from 2018-01-03T00:53:05.801.202.000 to 2018-01-03T00:53:05.801.491.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.237625

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      10 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.4766242 to 33.4769132
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-01-03T00:53:16.039.116.000 to 2018-01-03T00:53:16.039.405.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-01-03T00:52:45.614.146.750
   Max SL corrected packet time: 2018-01-03T00:53:16.039.264.750
   Min SL corrected XO rel time: 1.6142886
   Max SL corrected XO rel time: 32.0394062
   TM values from -28 to 18 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-01-03T00:52:45.614.146.750 (XO: 1.6142886, t=32.4°C)  Rank: 00008
   TM values: Max =    16  Min =   -24  Ave =    -4.46  Std =     6.27
   Min TM value:-24 at an offset of 128.263 usec -> 2018-01-03T00:52:45.614.275.013 (XO: 1.6144169)
   Max TM value:16 at an offset of 127.376 usec -> 2018-01-03T00:52:45.614.274.125 (XO: 1.6144160)
   Max dTM/dt [1/us]::3.34e+02 at an offset of 159.563 usec -> 2018-01-03T00:52:45.614.306.313 (XO: 1.6144482)
   Min dTM/dt [1/us]::-3.33e+02 at an offset of 159.638 usec -> 2018-01-03T00:52:45.614.306.388 (XO: 1.6144483)

 - Block number: 2 starting at 2018-01-03T00:53:05.801.061.750 (XO: 21.8012033, t=32.4°C)  Rank: 00009
   TM values: Max =    18  Min =   -26  Ave =    -4.36  Std =     7.41
   Min TM value:-26 at an offset of 176.676 usec -> 2018-01-03T00:53:05.801.238.425 (XO: 21.8013800)
   Max TM value:18 at an offset of 175.926 usec -> 2018-01-03T00:53:05.801.237.675 (XO: 21.8013792)
   Max dTM/dt [1/us]::3.66e+02 at an offset of 176.713 usec -> 2018-01-03T00:53:05.801.238.463 (XO: 21.8013800)
   Min dTM/dt [1/us]::-3.61e+02 at an offset of 175.963 usec -> 2018-01-03T00:53:05.801.237.713 (XO: 21.8013792)

 - Block number: 3 starting at 2018-01-03T00:53:16.038.975.750 (XO: 32.0391172, t=32.4°C)  Rank: 00010
   TM values: Max =    18  Min =   -28  Ave =    -4.40  Std =     8.21
   Min TM value:-28 at an offset of 162.626 usec -> 2018-01-03T00:53:16.039.138.375 (XO: 32.0392798)
   Max TM value:18 at an offset of 165.688 usec -> 2018-01-03T00:53:16.039.141.438 (XO: 32.0392828)
   Max dTM/dt [1/us]::3.87e+02 at an offset of 167.976 usec -> 2018-01-03T00:53:16.039.143.725 (XO: 32.0392851)
   Min dTM/dt [1/us]::-3.80e+02 at an offset of 167.901 usec -> 2018-01-03T00:53:16.039.143.650 (XO: 32.0392851)
