-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180102_054233_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 7.956074 to 31.853403
-each separately from 7s to 31s and from 175064us to 956363us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.218701 in 1 packets - #/No: 18/18
      16.677761 in 1 packets - #/No: 36/18
MIU packet time from 2018-01-02T05:42:43.956.074.000 to 2018-01-02T05:43:07.853.403.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.218701 in 1 packets - #/No: 18/18
      16.677761 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.4331279 to 32.4330315
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.9999678 in 1 packets - #/No: 18/18
      15.9999358 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      05 -> 48.5°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180102_054233_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.3891997 to 9.3894887
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.2186710  -- OK for SL --
   MIU time from 2018-01-02T05:42:43.956.074.000 to 2018-01-02T05:42:43.956.363.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.218701

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.6081597 to 16.6084487
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.6776950  -- OK for SL --
   MIU time from 2018-01-02T05:42:51.175.064.000 to 2018-01-02T05:42:51.175.353.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.677761

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.2861437 to 33.2864327
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-01-02T05:43:07.853.114.000 to 2018-01-02T05:43:07.853.403.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-01-02T05:42:43.955.933.750
   Max SL corrected packet time: 2018-01-02T05:43:07.853.262.750
   Min SL corrected XO rel time: 7.9560757
   Max SL corrected XO rel time: 31.8534046
   TM values from -19 to 10 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-01-02T05:42:43.955.933.750 (XO: 7.9560757, t=34.0°C)  Rank: 00003
   TM values: Max =    10  Min =   -19  Ave =    -4.55  Std =     4.32
   Min TM value:-19 at an offset of 107.438 usec -> 2018-01-02T05:42:43.956.041.187 (XO: 7.9561832)
   Max TM value:10 at an offset of 105.175 usec -> 2018-01-02T05:42:43.956.038.925 (XO: 7.9561809)
   Max dTM/dt [1/us]::1.92e+02 at an offset of 85.050 usec -> 2018-01-02T05:42:43.956.018.800 (XO: 7.9561608)
   Min dTM/dt [1/us]::-1.77e+02 at an offset of 47.988 usec -> 2018-01-02T05:42:43.955.981.737 (XO: 7.9561237)

 - Block number: 2 starting at 2018-01-02T05:42:51.174.923.750 (XO: 15.1750647, t=34.0°C)  Rank: 00002
   TM values: Max =     6  Min =   -16  Ave =    -4.66  Std =     3.40
   Min TM value:-16 at an offset of 94.088 usec -> 2018-01-02T05:42:51.175.017.837 (XO: 15.1751588)
   Max TM value:6 at an offset of 81.275 usec -> 2018-01-02T05:42:51.175.005.025 (XO: 15.1751460)
   Max dTM/dt [1/us]::1.37e+02 at an offset of 117.388 usec -> 2018-01-02T05:42:51.175.041.137 (XO: 15.1751821)
   Min dTM/dt [1/us]::-1.26e+02 at an offset of 21.063 usec -> 2018-01-02T05:42:51.174.944.812 (XO: 15.1750858)

 - Block number: 3 starting at 2018-01-02T05:43:07.852.973.750 (XO: 31.8531156, t=34.0°C)  Rank: 00002
   TM values: Max =     7  Min =   -15  Ave =    -4.46  Std =     3.14
   Min TM value:-15 at an offset of 133.326 usec -> 2018-01-02T05:43:07.853.107.075 (XO: 31.8532489)
   Max TM value:7 at an offset of 86.713 usec -> 2018-01-02T05:43:07.853.060.462 (XO: 31.8532023)
   Max dTM/dt [1/us]::1.46e+02 at an offset of 39.163 usec -> 2018-01-02T05:43:07.853.012.912 (XO: 31.8531548)
   Min dTM/dt [1/us]::-1.36e+02 at an offset of 73.725 usec -> 2018-01-02T05:43:07.853.047.475 (XO: 31.8531893)
