-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20180102_003842_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 2.825527 to 44.968451
-each separately from 2s to 44s and from 11001us to 968451us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.185185 in 1 packets - #/No: 18/18
      29.956872 in 1 packets - #/No: 36/18
MIU packet time from 2018-01-02T00:38:47.825.527.000 to 2018-01-02T00:39:29.968.451.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.185185 in 1 packets - #/No: 18/18
      29.956872 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.4407019 to 45.4405311
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.9999471 in 1 packets - #/No: 18/18
      28.9998821 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      07 -> 45.9°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      BE -> 32.4°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20180102_003842_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.2662271 to 4.2665161
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.1851350  -- OK for SL --
   MIU time from 2018-01-02T00:38:47.825.527.000 to 2018-01-02T00:38:47.825.816.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.185185

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4516511 to 16.4519401
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 29.9567510  -- OK for SL --
   MIU time from 2018-01-02T00:39:00.011.001.000 to 2018-01-02T00:39:00.011.290.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 29.956872

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 46.4086911 to 46.4089801
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2018-01-02T00:39:29.968.162.000 to 2018-01-02T00:39:29.968.451.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2018-01-02T00:38:47.825.386.750
   Max SL corrected packet time: 2018-01-02T00:39:29.968.310.750
   Min SL corrected XO rel time: 2.8255285
   Max SL corrected XO rel time: 44.9684544
   TM values from -16 to 6 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2018-01-02T00:38:47.825.386.750 (XO: 2.8255285, t=32.4°C)  Rank: 00002
   TM values: Max =     6  Min =   -16  Ave =    -4.44  Std =     2.92
   Min TM value:-16 at an offset of 241.251 usec -> 2018-01-02T00:38:47.825.628.000 (XO: 2.8257698)
   Max TM value:6 at an offset of 122.263 usec -> 2018-01-02T00:38:47.825.509.012 (XO: 2.8256508)
   Max dTM/dt [1/us]::1.67e+02 at an offset of 161.388 usec -> 2018-01-02T00:38:47.825.548.138 (XO: 2.8256899)
   Min dTM/dt [1/us]::-1.88e+02 at an offset of 241.213 usec -> 2018-01-02T00:38:47.825.627.963 (XO: 2.8257697)

 - Block number: 2 starting at 2018-01-02T00:39:00.010.860.750 (XO: 15.0110021, t=32.4°C)  Rank: 00001
   TM values: Max =     4  Min =   -14  Ave =    -4.47  Std =     2.26
   Min TM value:-14 at an offset of 134.138 usec -> 2018-01-02T00:39:00.010.994.888 (XO: 15.0111363)
   Max TM value:4 at an offset of 262.501 usec -> 2018-01-02T00:39:00.011.123.251 (XO: 15.0112646)
   Max dTM/dt [1/us]::1.32e+02 at an offset of 260.451 usec -> 2018-01-02T00:39:00.011.121.201 (XO: 15.0112626)
   Min dTM/dt [1/us]::-1.36e+02 at an offset of 260.526 usec -> 2018-01-02T00:39:00.011.121.276 (XO: 15.0112627)

 - Block number: 3 starting at 2018-01-02T00:39:29.968.021.750 (XO: 44.9681654, t=32.4°C)  Rank: 00002
   TM values: Max =     6  Min =   -14  Ave =    -4.40  Std =     2.79
   Min TM value:-14 at an offset of 115.675 usec -> 2018-01-02T00:39:29.968.137.425 (XO: 44.9682810)
   Max TM value:6 at an offset of 268.914 usec -> 2018-01-02T00:39:29.968.290.663 (XO: 44.9684343)
   Max dTM/dt [1/us]::1.54e+02 at an offset of 198.288 usec -> 2018-01-02T00:39:29.968.220.038 (XO: 44.9683637)
   Min dTM/dt [1/us]::-1.39e+02 at an offset of 196.038 usec -> 2018-01-02T00:39:29.968.217.788 (XO: 44.9683614)
