-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171223_205538_1227.dat, 18  packets found.
APID:
      1227 (SL) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 17 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Hz and MHz counters from 15.377934 to 15.378223
-each separately from 15s to 15s and from 377934us to 378223us.
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU packet time from 2017-12-23T20:55:56.377.934.000 to 2017-12-23T20:55:56.378.223.000
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.5190308 to 14.5190308
   Differences in neigboring packets: 
      0.0000000 in 17 packets (all packets)
Temperature FPGA: 
      0C -> 39.4°C in 18 packets (all packets)
Temperature XO: 
      BE -> 32.4°C in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 18 packets (all packets)
Number of external events: 
      1 in 18 packets (all packets)
Number of generated alerts: 
      0 in 18 packets (all packets)
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      1 in 18 packets (all packets)
Zero: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      3 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20171223_205538_1227.dat, 1 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1228 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.8969645 to 14.8972535
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-12-23T20:55:56.377.934.000 to 2017-12-23T20:55:56.378.223.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-12-23T20:55:56.377.663.818
   Max SL corrected packet time: 2017-12-23T20:55:56.377.952.818
   Min SL corrected XO rel time: 28.6980260
   Max SL corrected XO rel time: 28.6985827
   TM values from -23 to 14 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-12-23T20:55:56.377.663.818 (XO: 28.6980260, t=32.4°C)  Rank: 01228
   TM values: Max =    14  Min =   -23  Ave =    -4.29  Std =     5.18
   Min TM value:-23 at an offset of 312.853 usec -> 2017-12-23T20:55:56.377.976.670 (XO: 28.6983389)
   Max TM value:14 at an offset of 9.801 usec -> 2017-12-23T20:55:56.377.673.618 (XO: 28.6980358)
   Max dTM/dt [1/us]::1.15e+02 at an offset of 343.291 usec -> 2017-12-23T20:55:56.378.007.108 (XO: 28.6983693)
   Min dTM/dt [1/us]::-1.37e+02 at an offset of 514.478 usec -> 2017-12-23T20:55:56.378.178.296 (XO: 28.6985405)
