-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171222_160850_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.234570 to 30.279664
-each separately from 0s to 30s and from 15988us to 279664us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.781129 in 1 packets - #/No: 18/18
      15.263098 in 1 packets - #/No: 36/18
MIU packet time from 2017-12-22T16:08:53.234.570.000 to 2017-12-22T16:09:23.279.664.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.781129 in 1 packets - #/No: 18/18
      15.263098 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4388141 to 31.4386970
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999414 in 1 packets - #/No: 18/18
      14.9999415 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      05 -> 48.5°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171222_160850_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.6733843 to 1.6736733
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.7810710  -- OK for SL --
   MIU time from 2017-12-22T16:08:53.234.570.000 to 2017-12-22T16:08:53.234.859.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.781129

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4547443 to 16.4550333
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.2630390  -- OK for SL --
   MIU time from 2017-12-22T16:09:08.015.988.000 to 2017-12-22T16:09:08.016.277.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.263098

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.7180723 to 31.7183613
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-12-22T16:09:23.279.375.000 to 2017-12-22T16:09:23.279.664.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-12-22T16:08:53.234.429.750
   Max SL corrected packet time: 2017-12-22T16:09:23.279.523.750
   Min SL corrected XO rel time: 0.2345711
   Max SL corrected XO rel time: 30.2796654
   TM values from -21 to 11 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-12-22T16:08:53.234.429.750 (XO: 0.2345711, t=34.0°C)  Rank: 00002
   TM values: Max =     9  Min =   -18  Ave =    -4.47  Std =     3.70
   Min TM value:-18 at an offset of 45.888 usec -> 2017-12-22T16:08:53.234.475.637 (XO: 0.2346170)
   Max TM value:9 at an offset of 22.413 usec -> 2017-12-22T16:08:53.234.452.162 (XO: 0.2345935)
   Max dTM/dt [1/us]::1.87e+02 at an offset of 44.963 usec -> 2017-12-22T16:08:53.234.474.712 (XO: 0.2346161)
   Min dTM/dt [1/us]::-1.61e+02 at an offset of 120.150 usec -> 2017-12-22T16:08:53.234.549.900 (XO: 0.2346913)

 - Block number: 2 starting at 2017-12-22T16:09:08.015.847.750 (XO: 15.0159889, t=34.0°C)  Rank: 00002
   TM values: Max =    10  Min =   -18  Ave =    -4.42  Std =     4.23
   Min TM value:-18 at an offset of 58.738 usec -> 2017-12-22T16:09:08.015.906.487 (XO: 15.0160476)
   Max TM value:10 at an offset of 42.888 usec -> 2017-12-22T16:09:08.015.890.637 (XO: 15.0160318)
   Max dTM/dt [1/us]::1.58e+02 at an offset of 176.326 usec -> 2017-12-22T16:09:08.016.024.075 (XO: 15.0161652)
   Min dTM/dt [1/us]::-1.64e+02 at an offset of 101.000 usec -> 2017-12-22T16:09:08.015.948.750 (XO: 15.0160899)

 - Block number: 3 starting at 2017-12-22T16:09:23.279.234.750 (XO: 30.2793764, t=34.0°C)  Rank: 00002
   TM values: Max =    11  Min =   -21  Ave =    -4.41  Std =     3.82
   Min TM value:-21 at an offset of 112.538 usec -> 2017-12-22T16:09:23.279.347.287 (XO: 30.2794889)
   Max TM value:11 at an offset of 14.625 usec -> 2017-12-22T16:09:23.279.249.375 (XO: 30.2793910)
   Max dTM/dt [1/us]::1.69e+02 at an offset of 227.638 usec -> 2017-12-22T16:09:23.279.462.388 (XO: 30.2796040)
   Min dTM/dt [1/us]::-1.86e+02 at an offset of 215.488 usec -> 2017-12-22T16:09:23.279.450.238 (XO: 30.2795919)
