-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171220_204554_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 13.043606 to 33.258398
-each separately from 13s to 33s and from 32589us to 258398us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      1.988694 in 1 packets - #/No: 18/18
      18.225231 in 1 packets - #/No: 36/18
MIU packet time from 2017-12-20T20:46:10.043.606.000 to 2017-12-20T20:46:30.258.398.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      1.988694 in 1 packets - #/No: 18/18
      18.225231 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.4393674 to 34.4392969
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      1.9999929 in 1 packets - #/No: 18/18
      17.9999366 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171220_204554_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.4829742 to 14.4832632
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 1.9886870  -- OK for SL --
   MIU time from 2017-12-20T20:46:10.043.606.000 to 2017-12-20T20:46:10.043.895.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 1.988694

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4719502 to 16.4722392
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.2251670  -- OK for SL --
   MIU time from 2017-12-20T20:46:12.032.589.000 to 2017-12-20T20:46:12.032.878.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.225231

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.6974062 to 34.6976952
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-12-20T20:46:30.258.109.000 to 2017-12-20T20:46:30.258.398.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-12-20T20:46:10.043.465.750
   Max SL corrected packet time: 2017-12-20T20:46:30.258.257.750
   Min SL corrected XO rel time: 13.0436070
   Max SL corrected XO rel time: 33.2583992
   TM values from -27 to 19 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-12-20T20:46:10.043.465.750 (XO: 13.0436070, t=35.6°C)  Rank: 00009
   TM values: Max =    18  Min =   -26  Ave =    -4.43  Std =     7.51
   Min TM value:-26 at an offset of 123.113 usec -> 2017-12-20T20:46:10.043.588.862 (XO: 13.0437301)
   Max TM value:18 at an offset of 169.601 usec -> 2017-12-20T20:46:10.043.635.350 (XO: 13.0437766)
   Max dTM/dt [1/us]::3.43e+02 at an offset of 121.988 usec -> 2017-12-20T20:46:10.043.587.737 (XO: 13.0437289)
   Min dTM/dt [1/us]::-3.59e+02 at an offset of 169.813 usec -> 2017-12-20T20:46:10.043.635.563 (XO: 13.0437768)

 - Block number: 2 starting at 2017-12-20T20:46:12.032.448.750 (XO: 15.0325900, t=35.6°C)  Rank: 00009
   TM values: Max =    19  Min =   -27  Ave =    -4.52  Std =     7.26
   Min TM value:-27 at an offset of 129.625 usec -> 2017-12-20T20:46:12.032.578.375 (XO: 15.0327196)
   Max TM value:19 at an offset of 128.875 usec -> 2017-12-20T20:46:12.032.577.625 (XO: 15.0327189)
   Max dTM/dt [1/us]::3.75e+02 at an offset of 128.825 usec -> 2017-12-20T20:46:12.032.577.575 (XO: 15.0327188)
   Min dTM/dt [1/us]::-3.69e+02 at an offset of 131.088 usec -> 2017-12-20T20:46:12.032.579.837 (XO: 15.0327211)

 - Block number: 3 starting at 2017-12-20T20:46:30.257.968.750 (XO: 33.2581102, t=35.6°C)  Rank: 00007
   TM values: Max =    14  Min =   -23  Ave =    -4.54  Std =     5.29
   Min TM value:-23 at an offset of 136.013 usec -> 2017-12-20T20:46:30.258.104.762 (XO: 33.2582462)
   Max TM value:14 at an offset of 117.200 usec -> 2017-12-20T20:46:30.258.085.950 (XO: 33.2582274)
   Max dTM/dt [1/us]::2.98e+02 at an offset of 126.525 usec -> 2017-12-20T20:46:30.258.095.275 (XO: 33.2582368)
   Min dTM/dt [1/us]::-3.19e+02 at an offset of 126.613 usec -> 2017-12-20T20:46:30.258.095.362 (XO: 33.2582368)
