-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171216_002224_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.115292 to 33.543149
-each separately from 0s to 33s and from 115292us to 693236us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      18.577366 in 1 packets - #/No: 18/18
      14.849624 in 1 packets - #/No: 36/18
MIU packet time from 2017-12-16T00:22:27.115.292.000 to 2017-12-16T00:23:00.543.149.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      18.577366 in 1 packets - #/No: 18/18
      14.849624 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.9435793 to 34.9434528
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      17.9999309 in 1 packets - #/No: 18/18
      14.9999426 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      05 -> 48.5°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171216_002224_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.0588717 to 2.0591607
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.5772950  -- OK for SL --
   MIU time from 2017-12-16T00:22:27.115.292.000 to 2017-12-16T00:22:27.115.581.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.577366

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.6364557 to 20.6367447
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.8495670  -- OK for SL --
   MIU time from 2017-12-16T00:22:45.692.947.000 to 2017-12-16T00:22:45.693.236.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.849624

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.4863117 to 35.4866007
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-12-16T00:23:00.542.860.000 to 2017-12-16T00:23:00.543.149.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-12-16T00:22:27.115.151.750
   Max SL corrected packet time: 2017-12-16T00:23:00.543.008.750
   Min SL corrected XO rel time: 0.1152928
   Max SL corrected XO rel time: 33.5431500
   TM values from -17 to 8 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-12-16T00:22:27.115.151.750 (XO: 0.1152928, t=34.0°C)  Rank: 00002
   TM values: Max =     7  Min =   -16  Ave =    -4.44  Std =     4.31
   Min TM value:-16 at an offset of 86.925 usec -> 2017-12-16T00:22:27.115.238.675 (XO: 0.1153798)
   Max TM value:7 at an offset of 174.226 usec -> 2017-12-16T00:22:27.115.325.975 (XO: 0.1154671)
   Max dTM/dt [1/us]::1.86e+02 at an offset of 87.800 usec -> 2017-12-16T00:22:27.115.239.550 (XO: 0.1153806)
   Min dTM/dt [1/us]::-1.83e+02 at an offset of 53.163 usec -> 2017-12-16T00:22:27.115.204.912 (XO: 0.1153460)

 - Block number: 2 starting at 2017-12-16T00:22:45.692.806.750 (XO: 18.6929481, t=34.0°C)  Rank: 00002
   TM values: Max =     8  Min =   -17  Ave =    -4.45  Std =     3.13
   Min TM value:-17 at an offset of 119.238 usec -> 2017-12-16T00:22:45.692.925.987 (XO: 18.6930674)
   Max TM value:8 at an offset of 104.388 usec -> 2017-12-16T00:22:45.692.911.137 (XO: 18.6930525)
   Max dTM/dt [1/us]::1.31e+02 at an offset of 96.713 usec -> 2017-12-16T00:22:45.692.903.462 (XO: 18.6930448)
   Min dTM/dt [1/us]::-1.33e+02 at an offset of 104.600 usec -> 2017-12-16T00:22:45.692.911.350 (XO: 18.6930527)

 - Block number: 3 starting at 2017-12-16T00:23:00.542.719.750 (XO: 33.5428610, t=34.0°C)  Rank: 00002
   TM values: Max =     3  Min =   -13  Ave =    -4.46  Std =     2.17
   Min TM value:-13 at an offset of 37.675 usec -> 2017-12-16T00:23:00.542.757.425 (XO: 33.5428986)
   Max TM value:3 at an offset of 0.838 usec -> 2017-12-16T00:23:00.542.720.587 (XO: 33.5428618)
   Max dTM/dt [1/us]::9.20e+01 at an offset of 110.975 usec -> 2017-12-16T00:23:00.542.830.725 (XO: 33.5429719)
   Min dTM/dt [1/us]::-8.70e+01 at an offset of 10.888 usec -> 2017-12-16T00:23:00.542.730.637 (XO: 33.5428719)
