-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171215_102328_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 9.244682 to 37.461494
-each separately from 9s to 37s and from 244682us to 734978us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.489718 in 1 packets - #/No: 18/18
      13.726227 in 1 packets - #/No: 36/18
MIU packet time from 2017-12-15T10:23:40.244.682.000 to 2017-12-15T10:24:08.461.494.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.489718 in 1 packets - #/No: 18/18
      13.726227 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.4341617 to 38.4340561
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.9999472 in 2 packets - #/No: 18/18, 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171215_102328_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.6788444 to 10.6791334
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.4896630  -- OK for SL --
   MIU time from 2017-12-15T10:23:40.244.682.000 to 2017-12-15T10:23:40.244.971.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.489718

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.1687964 to 25.1690854
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.7261750  -- OK for SL --
   MIU time from 2017-12-15T10:23:54.734.689.000 to 2017-12-15T10:23:54.734.978.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.726227

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 38.8952604 to 38.8955494
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-12-15T10:24:08.461.205.000 to 2017-12-15T10:24:08.461.494.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-12-15T10:23:40.244.541.750
   Max SL corrected packet time: 2017-12-15T10:24:08.461.353.750
   Min SL corrected XO rel time: 9.2446836
   Max SL corrected XO rel time: 37.4614951
   TM values from -16 to 9 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-12-15T10:23:40.244.541.750 (XO: 9.2446836, t=34.0°C)  Rank: 00002
   TM values: Max =     8  Min =   -16  Ave =    -4.48  Std =     4.24
   Min TM value:-16 at an offset of 22.138 usec -> 2017-12-15T10:23:40.244.563.887 (XO: 9.2447058)
   Max TM value:8 at an offset of 103.275 usec -> 2017-12-15T10:23:40.244.645.025 (XO: 9.2447869)
   Max dTM/dt [1/us]::9.70e+01 at an offset of 99.975 usec -> 2017-12-15T10:23:40.244.641.725 (XO: 9.2447836)
   Min dTM/dt [1/us]::-9.10e+01 at an offset of 216.426 usec -> 2017-12-15T10:23:40.244.758.175 (XO: 9.2449001)

 - Block number: 2 starting at 2017-12-15T10:23:54.734.548.750 (XO: 23.7346903, t=34.0°C)  Rank: 00002
   TM values: Max =     9  Min =   -16  Ave =    -4.40  Std =     3.29
   Min TM value:-16 at an offset of 162.926 usec -> 2017-12-15T10:23:54.734.711.675 (XO: 23.7348532)
   Max TM value:9 at an offset of 154.363 usec -> 2017-12-15T10:23:54.734.703.113 (XO: 23.7348447)
   Max dTM/dt [1/us]::8.20e+01 at an offset of 154.313 usec -> 2017-12-15T10:23:54.734.703.063 (XO: 23.7348446)
   Min dTM/dt [1/us]::-7.90e+01 at an offset of 115.188 usec -> 2017-12-15T10:23:54.734.663.937 (XO: 23.7348055)

 - Block number: 3 starting at 2017-12-15T10:24:08.461.064.750 (XO: 37.4612061, t=34.0°C)  Rank: 00002
   TM values: Max =     3  Min =   -12  Ave =    -4.44  Std =     2.14
   Min TM value:-12 at an offset of 89.725 usec -> 2017-12-15T10:24:08.461.154.475 (XO: 37.4612958)
   Max TM value:3 at an offset of 17.700 usec -> 2017-12-15T10:24:08.461.082.450 (XO: 37.4612238)
   Max dTM/dt [1/us]::5.40e+01 at an offset of 1.950 usec -> 2017-12-15T10:24:08.461.066.700 (XO: 37.4612080)
   Min dTM/dt [1/us]::-5.90e+01 at an offset of 89.688 usec -> 2017-12-15T10:24:08.461.154.437 (XO: 37.4612957)
