-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171212_134657_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 12.358165 to 30.115978
-each separately from 12s to 30s and from 115689us to 358454us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      2.998090 in 1 packets - #/No: 18/18
      14.758856 in 1 packets - #/No: 36/18
MIU packet time from 2017-12-12T13:47:12.358.165.000 to 2017-12-12T13:47:30.115.978.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      2.998090 in 1 packets - #/No: 18/18
      14.758856 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.4401981 to 31.4401298
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      2.9999885 in 1 packets - #/No: 18/18
      14.9999432 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      C0 -> 35.6°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171212_134657_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.7983631 to 13.7986521
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 2.9980790  -- OK for SL --
   MIU time from 2017-12-12T13:47:12.358.165.000 to 2017-12-12T13:47:12.358.454.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 2.998090

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.7967311 to 16.7970201
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.7587990  -- OK for SL --
   MIU time from 2017-12-12T13:47:15.356.544.000 to 2017-12-12T13:47:15.356.833.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.758856

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.5558191 to 31.5561081
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-12-12T13:47:30.115.689.000 to 2017-12-12T13:47:30.115.978.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-12-12T13:47:12.358.024.750
   Max SL corrected packet time: 2017-12-12T13:47:30.115.837.750
   Min SL corrected XO rel time: 12.3581664
   Max SL corrected XO rel time: 30.1159788
   TM values from -19 to 9 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-12-12T13:47:12.358.024.750 (XO: 12.3581664, t=34.0°C)  Rank: 00004
   TM values: Max =     9  Min =   -18  Ave =    -4.44  Std =     5.07
   Min TM value:-18 at an offset of 152.413 usec -> 2017-12-12T13:47:12.358.177.163 (XO: 12.3583188)
   Max TM value:9 at an offset of 122.613 usec -> 2017-12-12T13:47:12.358.147.362 (XO: 12.3582890)
   Max dTM/dt [1/us]::1.03e+02 at an offset of 156.763 usec -> 2017-12-12T13:47:12.358.181.513 (XO: 12.3583231)
   Min dTM/dt [1/us]::-1.06e+02 at an offset of 222.138 usec -> 2017-12-12T13:47:12.358.246.888 (XO: 12.3583885)

 - Block number: 2 starting at 2017-12-12T13:47:15.356.403.750 (XO: 15.3565459, t=34.0°C)  Rank: 00004
   TM values: Max =     9  Min =   -19  Ave =    -4.35  Std =     5.42
   Min TM value:-19 at an offset of 92.563 usec -> 2017-12-12T13:47:15.356.496.312 (XO: 15.3566384)
   Max TM value:9 at an offset of 96.438 usec -> 2017-12-12T13:47:15.356.500.187 (XO: 15.3566423)
   Max dTM/dt [1/us]::1.27e+02 at an offset of 92.613 usec -> 2017-12-12T13:47:15.356.496.362 (XO: 15.3566385)
   Min dTM/dt [1/us]::-1.13e+02 at an offset of 182.351 usec -> 2017-12-12T13:47:15.356.586.100 (XO: 15.3567282)

 - Block number: 3 starting at 2017-12-12T13:47:30.115.548.750 (XO: 30.1156898, t=34.0°C)  Rank: 00002
   TM values: Max =     5  Min =   -14  Ave =    -4.35  Std =     3.38
   Min TM value:-14 at an offset of 59.525 usec -> 2017-12-12T13:47:30.115.608.275 (XO: 30.1157493)
   Max TM value:5 at an offset of 59.038 usec -> 2017-12-12T13:47:30.115.607.787 (XO: 30.1157488)
   Max dTM/dt [1/us]::7.70e+01 at an offset of 63.025 usec -> 2017-12-12T13:47:30.115.611.775 (XO: 30.1157528)
   Min dTM/dt [1/us]::-7.70e+01 at an offset of 193.088 usec -> 2017-12-12T13:47:30.115.741.838 (XO: 30.1158828)
