-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171206_092637_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.382658 to 30.091416
-each separately from 0s to 30s and from 91127us to 553070us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.169834 in 1 packets - #/No: 18/18
      14.538057 in 1 packets - #/No: 36/18
MIU packet time from 2017-12-06T09:26:40.382.658.000 to 2017-12-06T09:27:10.091.416.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.169834 in 1 packets - #/No: 18/18
      14.538057 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4355588 to 31.4354413
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999413 in 1 packets - #/No: 18/18
      14.9999412 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      06 -> 47.2°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171206_092637_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.8182168 to 1.8185058
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.1697750  -- OK for SL --
   MIU time from 2017-12-06T09:26:40.382.658.000 to 2017-12-06T09:26:40.382.947.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.169834

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.9882808 to 16.9885698
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.5379990  -- OK for SL --
   MIU time from 2017-12-06T09:26:55.552.781.000 to 2017-12-06T09:26:55.553.070.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.538057

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.5265688 to 31.5268578
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-12-06T09:27:10.091.127.000 to 2017-12-06T09:27:10.091.416.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-12-06T09:26:40.382.517.750
   Max SL corrected packet time: 2017-12-06T09:27:10.091.275.750
   Min SL corrected XO rel time: 0.3826595
   Max SL corrected XO rel time: 30.0914169
   TM values from -21 to 12 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-12-06T09:26:40.382.517.750 (XO: 0.3826595, t=34.0°C)  Rank: 00004
   TM values: Max =    12  Min =   -21  Ave =    -4.43  Std =     5.32
   Min TM value:-21 at an offset of 76.550 usec -> 2017-12-06T09:26:40.382.594.300 (XO: 0.3827360)
   Max TM value:12 at an offset of 81.688 usec -> 2017-12-06T09:26:40.382.599.437 (XO: 0.3827412)
   Max dTM/dt [1/us]::1.49e+02 at an offset of 76.988 usec -> 2017-12-06T09:26:40.382.594.737 (XO: 0.3827365)
   Min dTM/dt [1/us]::-1.41e+02 at an offset of 76.113 usec -> 2017-12-06T09:26:40.382.593.862 (XO: 0.3827356)

 - Block number: 2 starting at 2017-12-06T09:26:55.552.640.750 (XO: 15.5527829, t=34.0°C)  Rank: 00003
   TM values: Max =    12  Min =   -20  Ave =    -4.40  Std =     5.22
   Min TM value:-20 at an offset of 123.425 usec -> 2017-12-06T09:26:55.552.764.175 (XO: 15.5529063)
   Max TM value:12 at an offset of 148.613 usec -> 2017-12-06T09:26:55.552.789.363 (XO: 15.5529315)
   Max dTM/dt [1/us]::1.60e+02 at an offset of 143.801 usec -> 2017-12-06T09:26:55.552.784.550 (XO: 15.5529267)
   Min dTM/dt [1/us]::-1.62e+02 at an offset of 186.963 usec -> 2017-12-06T09:26:55.552.827.713 (XO: 15.5529698)

 - Block number: 3 starting at 2017-12-06T09:27:10.090.986.750 (XO: 30.0911279, t=34.0°C)  Rank: 00002
   TM values: Max =     9  Min =   -18  Ave =    -4.52  Std =     4.28
   Min TM value:-18 at an offset of 35.850 usec -> 2017-12-06T09:27:10.091.022.600 (XO: 30.0911637)
   Max TM value:9 at an offset of 300.189 usec -> 2017-12-06T09:27:10.091.286.938 (XO: 30.0914280)
   Max dTM/dt [1/us]::1.28e+02 at an offset of 115.250 usec -> 2017-12-06T09:27:10.091.102.000 (XO: 30.0912431)
   Min dTM/dt [1/us]::-1.30e+02 at an offset of 163.363 usec -> 2017-12-06T09:27:10.091.150.113 (XO: 30.0912912)
