-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171205_114509_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 2.632608 to 31.802608
-each separately from 2s to 31s and from 55536us to 802608us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.422639 in 1 packets - #/No: 18/18
      16.746494 in 1 packets - #/No: 36/18
MIU packet time from 2017-12-05T11:45:14.632.608.000 to 2017-12-05T11:45:43.802.608.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.422639 in 1 packets - #/No: 18/18
      16.746494 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.4334655 to 32.4333557
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      12.9999507 in 1 packets - #/No: 18/18
      15.9999395 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      06 -> 47.2°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171205_114509_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.0660731 to 4.0663621
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.4225910  -- OK for SL --
   MIU time from 2017-12-05T11:45:14.632.608.000 to 2017-12-05T11:45:14.632.897.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.422639

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4889531 to 16.4892421
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.7464310  -- OK for SL --
   MIU time from 2017-12-05T11:45:27.055.536.000 to 2017-12-05T11:45:27.055.825.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.746494

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.2356731 to 33.2359621
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-12-05T11:45:43.802.319.000 to 2017-12-05T11:45:43.802.608.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-12-05T11:45:14.632.467.750
   Max SL corrected packet time: 2017-12-05T11:45:43.802.467.750
   Min SL corrected XO rel time: 2.6326100
   Max SL corrected XO rel time: 31.8026094
   TM values from -23 to 14 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-12-05T11:45:14.632.467.750 (XO: 2.6326100, t=34.0°C)  Rank: 00003
   TM values: Max =    14  Min =   -23  Ave =    -4.44  Std =     5.89
   Min TM value:-23 at an offset of 149.913 usec -> 2017-12-05T11:45:14.632.617.663 (XO: 2.6327599)
   Max TM value:14 at an offset of 127.675 usec -> 2017-12-05T11:45:14.632.595.425 (XO: 2.6327377)
   Max dTM/dt [1/us]::1.48e+02 at an offset of 127.300 usec -> 2017-12-05T11:45:14.632.595.050 (XO: 2.6327373)
   Min dTM/dt [1/us]::-1.60e+02 at an offset of 127.725 usec -> 2017-12-05T11:45:14.632.595.475 (XO: 2.6327377)

 - Block number: 2 starting at 2017-12-05T11:45:27.055.395.750 (XO: 15.0555371, t=34.0°C)  Rank: 00002
   TM values: Max =    10  Min =   -18  Ave =    -4.56  Std =     3.79
   Min TM value:-18 at an offset of 102.363 usec -> 2017-12-05T11:45:27.055.498.112 (XO: 15.0556395)
   Max TM value:10 at an offset of 142.726 usec -> 2017-12-05T11:45:27.055.538.475 (XO: 15.0556798)
   Max dTM/dt [1/us]::1.07e+02 at an offset of 237.976 usec -> 2017-12-05T11:45:27.055.633.725 (XO: 15.0557751)
   Min dTM/dt [1/us]::-9.60e+01 at an offset of 238.851 usec -> 2017-12-05T11:45:27.055.634.600 (XO: 15.0557760)

 - Block number: 3 starting at 2017-12-05T11:45:43.802.178.750 (XO: 31.8023204, t=34.0°C)  Rank: 00002
   TM values: Max =     9  Min =   -19  Ave =    -4.63  Std =     3.87
   Min TM value:-19 at an offset of 136.351 usec -> 2017-12-05T11:45:43.802.315.100 (XO: 31.8024568)
   Max TM value:9 at an offset of 64.200 usec -> 2017-12-05T11:45:43.802.242.950 (XO: 31.8023846)
   Max dTM/dt [1/us]::1.22e+02 at an offset of 170.088 usec -> 2017-12-05T11:45:43.802.348.838 (XO: 31.8024905)
   Min dTM/dt [1/us]::-1.16e+02 at an offset of 135.513 usec -> 2017-12-05T11:45:43.802.314.263 (XO: 31.8024560)
