-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171205_101421_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 1.927635 to 30.980710
-each separately from 1s to 30s and from 200983us to 980710us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.273059 in 1 packets - #/No: 18/18
      15.779149 in 1 packets - #/No: 36/18
MIU packet time from 2017-12-05T10:14:25.927.635.000 to 2017-12-05T10:14:54.980.710.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.273059 in 1 packets - #/No: 18/18
      15.779149 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.4368402 to 31.4367268
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      13.9999453 in 1 packets - #/No: 18/18
      14.9999413 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171205_101421_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.3644732 to 3.3647622
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.2730070  -- OK for SL --
   MIU time from 2017-12-05T10:14:25.927.635.000 to 2017-12-05T10:14:25.927.924.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.273059

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.6377692 to 16.6380582
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.7790870  -- OK for SL --
   MIU time from 2017-12-05T10:14:39.200.983.000 to 2017-12-05T10:14:39.201.272.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.779149

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.4171452 to 32.4174342
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-12-05T10:14:54.980.421.000 to 2017-12-05T10:14:54.980.710.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-12-05T10:14:25.927.494.750
   Max SL corrected packet time: 2017-12-05T10:14:54.980.569.750
   Min SL corrected XO rel time: 1.9276366
   Max SL corrected XO rel time: 30.9807112
   TM values from -18 to 9 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-12-05T10:14:25.927.494.750 (XO: 1.9276366, t=34.0°C)  Rank: 00002
   TM values: Max =     9  Min =   -18  Ave =    -4.39  Std =     4.26
   Min TM value:-18 at an offset of 142.463 usec -> 2017-12-05T10:14:25.927.637.213 (XO: 1.9277791)
   Max TM value:9 at an offset of 213.576 usec -> 2017-12-05T10:14:25.927.708.325 (XO: 1.9278502)
   Max dTM/dt [1/us]::1.15e+02 at an offset of 142.963 usec -> 2017-12-05T10:14:25.927.637.713 (XO: 1.9277796)
   Min dTM/dt [1/us]::-1.02e+02 at an offset of 242.926 usec -> 2017-12-05T10:14:25.927.737.675 (XO: 1.9278795)

 - Block number: 2 starting at 2017-12-05T10:14:39.200.842.750 (XO: 15.2009845, t=34.0°C)  Rank: 00002
   TM values: Max =     9  Min =   -18  Ave =    -4.39  Std =     4.32
   Min TM value:-18 at an offset of 134.126 usec -> 2017-12-05T10:14:39.200.976.875 (XO: 15.2011186)
   Max TM value:9 at an offset of 172.801 usec -> 2017-12-05T10:14:39.201.015.550 (XO: 15.2011573)
   Max dTM/dt [1/us]::9.80e+01 at an offset of 69.225 usec -> 2017-12-05T10:14:39.200.911.975 (XO: 15.2010537)
   Min dTM/dt [1/us]::-1.01e+02 at an offset of 42.225 usec -> 2017-12-05T10:14:39.200.884.975 (XO: 15.2010267)

 - Block number: 3 starting at 2017-12-05T10:14:54.980.280.750 (XO: 30.9804222, t=34.0°C)  Rank: 00003
   TM values: Max =     8  Min =   -18  Ave =    -4.46  Std =     4.29
   Min TM value:-18 at an offset of 99.988 usec -> 2017-12-05T10:14:54.980.380.737 (XO: 30.9805222)
   Max TM value:8 at an offset of 78.075 usec -> 2017-12-05T10:14:54.980.358.825 (XO: 30.9805003)
   Max dTM/dt [1/us]::1.27e+02 at an offset of 100.038 usec -> 2017-12-05T10:14:54.980.380.787 (XO: 30.9805223)
   Min dTM/dt [1/us]::-1.15e+02 at an offset of 131.413 usec -> 2017-12-05T10:14:54.980.412.163 (XO: 30.9805536)
