-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171130_170346_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.006288 to 40.946173
-each separately from 0s to 40s and from 6288us to 946173us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      24.895630 in 1 packets - #/No: 18/18
      16.043388 in 1 packets - #/No: 36/18
MIU packet time from 2017-11-30T17:03:49.006.288.000 to 2017-11-30T17:04:29.946.173.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      24.895630 in 1 packets - #/No: 18/18
      16.043388 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4403863 to 41.4402328
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      23.9999079 in 1 packets - #/No: 18/18
      15.9999386 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      07 -> 45.9°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171130_170346_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4466750 to 1.4469640
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.8955350  -- OK for SL --
   MIU time from 2017-11-30T17:03:49.006.288.000 to 2017-11-30T17:03:49.006.577.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.895630

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.3424990 to 26.3427880
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.0433270  -- OK for SL --
   MIU time from 2017-11-30T17:04:13.902.207.000 to 2017-11-30T17:04:13.902.496.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.043388

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 42.3861150 to 42.3864040
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-11-30T17:04:29.945.884.000 to 2017-11-30T17:04:29.946.173.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-11-30T17:03:49.006.147.750
   Max SL corrected packet time: 2017-11-30T17:04:29.946.032.750
   Min SL corrected XO rel time: 0.0062887
   Max SL corrected XO rel time: 40.9461749
   TM values from -20 to 12 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-11-30T17:03:49.006.147.750 (XO: 0.0062887, t=34.0°C)  Rank: 00002
   TM values: Max =     8  Min =   -20  Ave =    -4.45  Std =     4.05
   Min TM value:-20 at an offset of 111.088 usec -> 2017-11-30T17:03:49.006.258.837 (XO: 0.0063998)
   Max TM value:8 at an offset of 25.000 usec -> 2017-11-30T17:03:49.006.172.750 (XO: 0.0063137)
   Max dTM/dt [1/us]::1.80e+02 at an offset of 97.888 usec -> 2017-11-30T17:03:49.006.245.637 (XO: 0.0063866)
   Min dTM/dt [1/us]::-1.88e+02 at an offset of 97.275 usec -> 2017-11-30T17:03:49.006.245.025 (XO: 0.0063860)

 - Block number: 2 starting at 2017-11-30T17:04:13.902.066.750 (XO: 24.9022083, t=34.0°C)  Rank: 00003
   TM values: Max =    12  Min =   -20  Ave =    -4.35  Std =     4.23
   Min TM value:-20 at an offset of 41.000 usec -> 2017-11-30T17:04:13.902.107.750 (XO: 24.9022493)
   Max TM value:12 at an offset of 250.338 usec -> 2017-11-30T17:04:13.902.317.088 (XO: 24.9024587)
   Max dTM/dt [1/us]::1.82e+02 at an offset of 128.013 usec -> 2017-11-30T17:04:13.902.194.762 (XO: 24.9023363)
   Min dTM/dt [1/us]::-2.07e+02 at an offset of 249.538 usec -> 2017-11-30T17:04:13.902.316.288 (XO: 24.9024579)

 - Block number: 3 starting at 2017-11-30T17:04:29.945.743.750 (XO: 40.9458859, t=34.0°C)  Rank: 00004
   TM values: Max =    12  Min =   -20  Ave =    -4.40  Std =     4.30
   Min TM value:-20 at an offset of 47.925 usec -> 2017-11-30T17:04:29.945.791.675 (XO: 40.9459338)
   Max TM value:12 at an offset of 49.225 usec -> 2017-11-30T17:04:29.945.792.975 (XO: 40.9459351)
   Max dTM/dt [1/us]::2.61e+02 at an offset of 49.338 usec -> 2017-11-30T17:04:29.945.793.087 (XO: 40.9459352)
   Min dTM/dt [1/us]::-2.74e+02 at an offset of 49.263 usec -> 2017-11-30T17:04:29.945.793.012 (XO: 40.9459352)
