-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171128_013515_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.164349 to 30.309236
-each separately from 0s to 30s and from 10999us to 309236us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.846361 in 1 packets - #/No: 18/18
      15.297659 in 1 packets - #/No: 36/18
MIU packet time from 2017-11-28T01:35:18.164.349.000 to 2017-11-28T01:35:48.309.236.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.846361 in 1 packets - #/No: 18/18
      15.297659 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4341966 to 31.4340793
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999413 in 1 packets - #/No: 18/18
      14.9999414 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      05 -> 48.5°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      BF -> 34.0°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      C0 -> 35.6°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171128_013515_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.5985463 to 1.5988353
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.8463030  -- OK for SL --
   MIU time from 2017-11-28T01:35:18.164.349.000 to 2017-11-28T01:35:18.164.638.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.846361

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4451383 to 16.4454273
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.2975990  -- OK for SL --
   MIU time from 2017-11-28T01:35:33.010.999.000 to 2017-11-28T01:35:33.011.288.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.297659

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.7430263 to 31.7433153
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-11-28T01:35:48.308.947.000 to 2017-11-28T01:35:48.309.236.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-11-28T01:35:18.164.208.750
   Max SL corrected packet time: 2017-11-28T01:35:48.309.095.750
   Min SL corrected XO rel time: 0.1643503
   Max SL corrected XO rel time: 30.3092372
   TM values from -16 to 7 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-11-28T01:35:18.164.208.750 (XO: 0.1643503, t=34.0°C)  Rank: 00002
   TM values: Max =     6  Min =   -16  Ave =    -4.36  Std =     2.74
   Min TM value:-16 at an offset of 137.601 usec -> 2017-11-28T01:35:18.164.346.350 (XO: 0.1644879)
   Max TM value:6 at an offset of 76.313 usec -> 2017-11-28T01:35:18.164.285.062 (XO: 0.1644267)
   Max dTM/dt [1/us]::1.29e+02 at an offset of 259.151 usec -> 2017-11-28T01:35:18.164.467.901 (XO: 0.1646095)
   Min dTM/dt [1/us]::-1.52e+02 at an offset of 118.963 usec -> 2017-11-28T01:35:18.164.327.712 (XO: 0.1644693)

 - Block number: 2 starting at 2017-11-28T01:35:33.010.858.750 (XO: 15.0110004, t=34.0°C)  Rank: 00001
   TM values: Max =     2  Min =   -13  Ave =    -4.51  Std =     2.01
   Min TM value:-13 at an offset of 66.713 usec -> 2017-11-28T01:35:33.010.925.462 (XO: 15.0110672)
   Max TM value:2 at an offset of 7.013 usec -> 2017-11-28T01:35:33.010.865.762 (XO: 15.0110075)
   Max dTM/dt [1/us]::9.50e+01 at an offset of 66.750 usec -> 2017-11-28T01:35:33.010.925.500 (XO: 15.0110672)
   Min dTM/dt [1/us]::-8.90e+01 at an offset of 103.188 usec -> 2017-11-28T01:35:33.010.961.937 (XO: 15.0111036)

 - Block number: 3 starting at 2017-11-28T01:35:48.308.806.750 (XO: 30.3089482, t=34.0°C)  Rank: 00002
   TM values: Max =     7  Min =   -14  Ave =    -4.44  Std =     2.62
   Min TM value:-14 at an offset of 0.613 usec -> 2017-11-28T01:35:48.308.807.362 (XO: 30.3089488)
   Max TM value:7 at an offset of 112.800 usec -> 2017-11-28T01:35:48.308.919.550 (XO: 30.3090610)
   Max dTM/dt [1/us]::1.41e+02 at an offset of 261.726 usec -> 2017-11-28T01:35:48.309.068.476 (XO: 30.3092099)
   Min dTM/dt [1/us]::-1.27e+02 at an offset of 243.201 usec -> 2017-11-28T01:35:48.309.049.950 (XO: 30.3091914)
