-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171126_111413_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.286225 to 31.261180
-each separately from 0s to 31s and from 20507us to 286514us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.733993 in 1 packets - #/No: 18/18
      16.240095 in 1 packets - #/No: 36/18
MIU packet time from 2017-11-26T11:14:16.286.225.000 to 2017-11-26T11:14:47.261.180.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.733993 in 1 packets - #/No: 18/18
      16.240095 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4333526 to 32.4332311
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999411 in 1 packets - #/No: 18/18
      15.9999374 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      06 -> 47.2°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171126_111413_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.7195780 to 1.7198670
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.7339350  -- OK for SL --
   MIU time from 2017-11-26T11:14:16.286.225.000 to 2017-11-26T11:14:16.286.514.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.733993

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4538020 to 16.4540910
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.2400310  -- OK for SL --
   MIU time from 2017-11-26T11:14:31.020.507.000 to 2017-11-26T11:14:31.020.796.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.240095

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.6941220 to 32.6944110
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-11-26T11:14:47.260.891.000 to 2017-11-26T11:14:47.261.180.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-11-26T11:14:16.286.084.750
   Max SL corrected packet time: 2017-11-26T11:14:47.261.039.750
   Min SL corrected XO rel time: 0.2862265
   Max SL corrected XO rel time: 31.2611809
   TM values from -21 to 12 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-11-26T11:14:16.286.084.750 (XO: 0.2862265, t=35.6°C)  Rank: 00003
   TM values: Max =    12  Min =   -21  Ave =    -4.50  Std =     4.74
   Min TM value:-21 at an offset of 94.063 usec -> 2017-11-26T11:14:16.286.178.812 (XO: 0.2863206)
   Max TM value:12 at an offset of 108.475 usec -> 2017-11-26T11:14:16.286.193.225 (XO: 0.2863350)
   Max dTM/dt [1/us]::1.29e+02 at an offset of 172.013 usec -> 2017-11-26T11:14:16.286.256.763 (XO: 0.2863985)
   Min dTM/dt [1/us]::-1.26e+02 at an offset of 108.913 usec -> 2017-11-26T11:14:16.286.193.662 (XO: 0.2863354)

 - Block number: 2 starting at 2017-11-26T11:14:31.020.366.750 (XO: 15.0205084, t=35.6°C)  Rank: 00002
   TM values: Max =    10  Min =   -18  Ave =    -4.47  Std =     4.41
   Min TM value:-18 at an offset of 43.625 usec -> 2017-11-26T11:14:31.020.410.375 (XO: 15.0205520)
   Max TM value:10 at an offset of 102.238 usec -> 2017-11-26T11:14:31.020.468.987 (XO: 15.0206106)
   Max dTM/dt [1/us]::1.22e+02 at an offset of 103.375 usec -> 2017-11-26T11:14:31.020.470.125 (XO: 15.0206118)
   Min dTM/dt [1/us]::-1.22e+02 at an offset of 102.675 usec -> 2017-11-26T11:14:31.020.469.425 (XO: 15.0206111)

 - Block number: 3 starting at 2017-11-26T11:14:47.260.750.750 (XO: 31.2608919, t=35.6°C)  Rank: 00002
   TM values: Max =     9  Min =   -19  Ave =    -4.47  Std =     3.92
   Min TM value:-19 at an offset of 109.113 usec -> 2017-11-26T11:14:47.260.859.862 (XO: 31.2610011)
   Max TM value:9 at an offset of 62.588 usec -> 2017-11-26T11:14:47.260.813.337 (XO: 31.2609545)
   Max dTM/dt [1/us]::1.13e+02 at an offset of 147.513 usec -> 2017-11-26T11:14:47.260.898.263 (XO: 31.2610395)
   Min dTM/dt [1/us]::-1.10e+02 at an offset of 108.675 usec -> 2017-11-26T11:14:47.260.859.425 (XO: 31.2610006)
