-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171125_015547_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 6.745895 to 35.278213
-each separately from 6s to 35s and from 277924us to 746184us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.641184 in 1 packets - #/No: 18/18
      19.890267 in 1 packets - #/No: 36/18
MIU packet time from 2017-11-25T01:55:56.745.895.000 to 2017-11-25T01:56:25.278.213.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.641184 in 1 packets - #/No: 18/18
      19.890267 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.9410129 to 36.9409021
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      8.9999655 in 1 packets - #/No: 18/18
      19.9999237 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171125_015547_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.6869063 to 8.6871953
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.6411510  -- OK for SL --
   MIU time from 2017-11-25T01:55:56.745.895.000 to 2017-11-25T01:55:56.746.184.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.641184

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.3283463 to 17.3286353
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.8901910  -- OK for SL --
   MIU time from 2017-11-25T01:56:05.387.368.000 to 2017-11-25T01:56:05.387.657.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.890267

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.2188263 to 37.2191153
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-11-25T01:56:25.277.924.000 to 2017-11-25T01:56:25.278.213.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-11-25T01:55:56.745.754.750
   Max SL corrected packet time: 2017-11-25T01:56:25.278.072.750
   Min SL corrected XO rel time: 6.7458963
   Max SL corrected XO rel time: 35.2782143
   TM values from -22 to 13 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-11-25T01:55:56.745.754.750 (XO: 6.7458963, t=34.0°C)  Rank: 00004
   TM values: Max =    13  Min =   -21  Ave =    -4.45  Std =     4.38
   Min TM value:-21 at an offset of 102.225 usec -> 2017-11-25T01:55:56.745.856.975 (XO: 6.7459985)
   Max TM value:13 at an offset of 94.000 usec -> 2017-11-25T01:55:56.745.848.750 (XO: 6.7459903)
   Max dTM/dt [1/us]::2.47e+02 at an offset of 102.263 usec -> 2017-11-25T01:55:56.745.857.012 (XO: 6.7459986)
   Min dTM/dt [1/us]::-2.39e+02 at an offset of 108.863 usec -> 2017-11-25T01:55:56.745.863.612 (XO: 6.7460052)

 - Block number: 2 starting at 2017-11-25T01:56:05.387.227.750 (XO: 15.3873694, t=34.0°C)  Rank: 00003
   TM values: Max =    10  Min =   -21  Ave =    -4.51  Std =     3.81
   Min TM value:-21 at an offset of 123.738 usec -> 2017-11-25T01:56:05.387.351.487 (XO: 15.3874931)
   Max TM value:10 at an offset of 122.988 usec -> 2017-11-25T01:56:05.387.350.737 (XO: 15.3874924)
   Max dTM/dt [1/us]::1.88e+02 at an offset of 123.775 usec -> 2017-11-25T01:56:05.387.351.525 (XO: 15.3874931)
   Min dTM/dt [1/us]::-1.87e+02 at an offset of 197.426 usec -> 2017-11-25T01:56:05.387.425.175 (XO: 15.3875668)

 - Block number: 3 starting at 2017-11-25T01:56:25.277.783.750 (XO: 35.2779253, t=34.0°C)  Rank: 00003
   TM values: Max =    12  Min =   -22  Ave =    -4.44  Std =     4.75
   Min TM value:-22 at an offset of 229.288 usec -> 2017-11-25T01:56:25.278.013.038 (XO: 35.2781545)
   Max TM value:12 at an offset of 196.688 usec -> 2017-11-25T01:56:25.277.980.438 (XO: 35.2781219)
   Max dTM/dt [1/us]::2.04e+02 at an offset of 108.088 usec -> 2017-11-25T01:56:25.277.891.837 (XO: 35.2780333)
   Min dTM/dt [1/us]::-2.19e+02 at an offset of 108.838 usec -> 2017-11-25T01:56:25.277.892.587 (XO: 35.2780341)
