-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171120_210227_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 8.621255 to 30.789419
-each separately from 8s to 30s and from 464085us to 789419us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.842541 in 1 packets - #/No: 18/18
      1.324756 in 1 packets - #/No: 36/18
MIU packet time from 2017-11-20T21:02:38.621.255.000 to 2017-11-20T21:03:00.789.419.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.842541 in 1 packets - #/No: 18/18
      1.324756 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.4346315 to 31.4345492
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      20.9999215 in 1 packets - #/No: 18/18
      0.9999962 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171120_210227_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.0558858 to 10.0561748
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.8424630  -- OK for SL --
   MIU time from 2017-11-20T21:02:38.621.255.000 to 2017-11-20T21:02:38.621.544.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.842541

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.8986378 to 30.8989268
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 1.3247510  -- OK for SL --
   MIU time from 2017-11-20T21:02:59.464.085.000 to 2017-11-20T21:02:59.464.374.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 1.324756

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.2236778 to 32.2239668
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-11-20T21:03:00.789.130.000 to 2017-11-20T21:03:00.789.419.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-11-20T21:02:38.621.114.750
   Max SL corrected packet time: 2017-11-20T21:03:00.789.278.750
   Min SL corrected XO rel time: 8.6212567
   Max SL corrected XO rel time: 30.7894206
   TM values from -25 to 19 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-11-20T21:02:38.621.114.750 (XO: 8.6212567, t=35.6°C)  Rank: 00005
   TM values: Max =    19  Min =   -25  Ave =    -4.56  Std =     6.19
   Min TM value:-25 at an offset of 166.838 usec -> 2017-11-20T21:02:38.621.281.588 (XO: 8.6214235)
   Max TM value:19 at an offset of 112.875 usec -> 2017-11-20T21:02:38.621.227.625 (XO: 8.6213695)
   Max dTM/dt [1/us]::3.27e+02 at an offset of 112.838 usec -> 2017-11-20T21:02:38.621.227.587 (XO: 8.6213695)
   Min dTM/dt [1/us]::-3.24e+02 at an offset of 117.788 usec -> 2017-11-20T21:02:38.621.232.537 (XO: 8.6213744)

 - Block number: 2 starting at 2017-11-20T21:02:59.463.944.750 (XO: 29.4640866, t=35.6°C)  Rank: 00005
   TM values: Max =    14  Min =   -23  Ave =    -4.55  Std =     5.34
   Min TM value:-23 at an offset of 131.601 usec -> 2017-11-20T21:02:59.464.076.350 (XO: 29.4642182)
   Max TM value:14 at an offset of 110.988 usec -> 2017-11-20T21:02:59.464.055.737 (XO: 29.4641976)
   Max dTM/dt [1/us]::2.89e+02 at an offset of 124.400 usec -> 2017-11-20T21:02:59.464.069.150 (XO: 29.4642110)
   Min dTM/dt [1/us]::-3.05e+02 at an offset of 131.563 usec -> 2017-11-20T21:02:59.464.076.312 (XO: 29.4642181)

 - Block number: 3 starting at 2017-11-20T21:03:00.788.989.750 (XO: 30.7891316, t=35.6°C)  Rank: 00006
   TM values: Max =    16  Min =   -25  Ave =    -4.58  Std =     7.00
   Min TM value:-25 at an offset of 106.350 usec -> 2017-11-20T21:03:00.789.096.100 (XO: 30.7892379)
   Max TM value:16 at an offset of 91.963 usec -> 2017-11-20T21:03:00.789.081.712 (XO: 30.7892236)
   Max dTM/dt [1/us]::3.36e+02 at an offset of 108.413 usec -> 2017-11-20T21:03:00.789.098.162 (XO: 30.7892400)
   Min dTM/dt [1/us]::-3.21e+02 at an offset of 119.938 usec -> 2017-11-20T21:03:00.789.109.687 (XO: 30.7892515)
