-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171118_150355_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 5.809457 to 30.022290
-each separately from 5s to 30s and from 10998us to 809746us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      9.201252 in 1 packets - #/No: 18/18
      15.010714 in 1 packets - #/No: 36/18
MIU packet time from 2017-11-18T15:04:03.809.457.000 to 2017-11-18T15:04:28.022.290.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      9.201252 in 1 packets - #/No: 18/18
      15.010714 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.4371629 to 31.4370648
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      9.9999607 in 1 packets - #/No: 18/18
      14.9999412 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      05 -> 48.5°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171118_150355_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.2466187 to 7.2469077
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.2012150  -- OK for SL --
   MIU time from 2017-11-18T15:04:03.809.457.000 to 2017-11-18T15:04:03.809.746.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.201252

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.4481227 to 16.4484117
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0106550  -- OK for SL --
   MIU time from 2017-11-18T15:04:13.010.998.000 to 2017-11-18T15:04:13.011.287.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010714

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.4590667 to 31.4593557
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-11-18T15:04:28.022.001.000 to 2017-11-18T15:04:28.022.290.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-11-18T15:04:03.809.316.750
   Max SL corrected packet time: 2017-11-18T15:04:28.022.149.750
   Min SL corrected XO rel time: 5.8094590
   Max SL corrected XO rel time: 30.0222910
   TM values from -19 to 9 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-11-18T15:04:03.809.316.750 (XO: 5.8094590, t=35.6°C)  Rank: 00002
   TM values: Max =     9  Min =   -19  Ave =    -4.49  Std =     4.12
   Min TM value:-19 at an offset of 75.600 usec -> 2017-11-18T15:04:03.809.392.350 (XO: 5.8095346)
   Max TM value:9 at an offset of 36.438 usec -> 2017-11-18T15:04:03.809.353.187 (XO: 5.8094954)
   Max dTM/dt [1/us]::1.21e+02 at an offset of 73.625 usec -> 2017-11-18T15:04:03.809.390.375 (XO: 5.8095326)
   Min dTM/dt [1/us]::-1.34e+02 at an offset of 23.313 usec -> 2017-11-18T15:04:03.809.340.062 (XO: 5.8094823)

 - Block number: 2 starting at 2017-11-18T15:04:13.010.857.750 (XO: 15.0109991, t=35.6°C)  Rank: 00001
   TM values: Max =     5  Min =   -14  Ave =    -4.49  Std =     2.68
   Min TM value:-14 at an offset of 87.763 usec -> 2017-11-18T15:04:13.010.945.512 (XO: 15.0110869)
   Max TM value:5 at an offset of 97.713 usec -> 2017-11-18T15:04:13.010.955.462 (XO: 15.0110969)
   Max dTM/dt [1/us]::1.17e+02 at an offset of 234.588 usec -> 2017-11-18T15:04:13.011.092.338 (XO: 15.0112337)
   Min dTM/dt [1/us]::-1.07e+02 at an offset of 107.925 usec -> 2017-11-18T15:04:13.010.965.675 (XO: 15.0111071)

 - Block number: 3 starting at 2017-11-18T15:04:28.021.860.750 (XO: 30.0220020, t=35.6°C)  Rank: 00001
   TM values: Max =     6  Min =   -15  Ave =    -4.52  Std =     2.81
   Min TM value:-15 at an offset of 77.513 usec -> 2017-11-18T15:04:28.021.938.262 (XO: 30.0220795)
   Max TM value:6 at an offset of 160.551 usec -> 2017-11-18T15:04:28.022.021.300 (XO: 30.0221625)
   Max dTM/dt [1/us]::1.29e+02 at an offset of 245.126 usec -> 2017-11-18T15:04:28.022.105.875 (XO: 30.0222471)
   Min dTM/dt [1/us]::-1.08e+02 at an offset of 282.276 usec -> 2017-11-18T15:04:28.022.143.026 (XO: 30.0222843)
