-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171114_184507_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 10.117132 to 43.733767
-each separately from 10s to 43s and from 117132us to 733767us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      10.047971 in 1 packets - #/No: 18/18
      23.567797 in 1 packets - #/No: 36/18
MIU packet time from 2017-11-14T18:45:20.117.132.000 to 2017-11-14T18:45:53.733.767.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      10.047971 in 1 packets - #/No: 18/18
      23.567797 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.4377294 to 44.4376085
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      9.9999633 in 1 packets - #/No: 18/18
      22.9999158 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171114_184507_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.5548619 to 11.5551509
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.0479350  -- OK for SL --
   MIU time from 2017-11-14T18:45:20.117.132.000 to 2017-11-14T18:45:20.117.421.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.047971

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      16 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.6030859 to 21.6033749
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.5677110  -- OK for SL --
   MIU time from 2017-11-14T18:45:30.165.392.000 to 2017-11-14T18:45:30.165.681.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.567797

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      9 in 18 packets (all packets)  -- OK --
   10 MHz counter from 45.1710859 to 45.1713749
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-11-14T18:45:53.733.478.000 to 2017-11-14T18:45:53.733.767.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-11-14T18:45:20.116.991.750
   Max SL corrected packet time: 2017-11-14T18:45:53.733.626.750
   Min SL corrected XO rel time: 10.1171329
   Max SL corrected XO rel time: 43.7337691
   TM values from -38 to 31 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-11-14T18:45:20.116.991.750 (XO: 10.1171329, t=35.6°C)  Rank: 00009
   TM values: Max =    26  Min =   -35  Ave =    -4.53  Std =     8.05
   Min TM value:-35 at an offset of 123.700 usec -> 2017-11-14T18:45:20.117.115.450 (XO: 10.1172566)
   Max TM value:26 at an offset of 124.163 usec -> 2017-11-14T18:45:20.117.115.912 (XO: 10.1172571)
   Max dTM/dt [1/us]::4.57e+02 at an offset of 123.725 usec -> 2017-11-14T18:45:20.117.115.475 (XO: 10.1172566)
   Min dTM/dt [1/us]::-4.92e+02 at an offset of 124.338 usec -> 2017-11-14T18:45:20.117.116.087 (XO: 10.1172573)

 - Block number: 2 starting at 2017-11-14T18:45:30.165.251.750 (XO: 20.1653938, t=35.6°C)  Rank: 00016
   TM values: Max =    31  Min =   -38  Ave =    -4.52  Std =    10.29
   Min TM value:-38 at an offset of 81.688 usec -> 2017-11-14T18:45:30.165.333.437 (XO: 20.1654755)
   Max TM value:31 at an offset of 104.000 usec -> 2017-11-14T18:45:30.165.355.750 (XO: 20.1654978)
   Max dTM/dt [1/us]::5.69e+02 at an offset of 103.563 usec -> 2017-11-14T18:45:30.165.355.312 (XO: 20.1654974)
   Min dTM/dt [1/us]::-5.45e+02 at an offset of 102.675 usec -> 2017-11-14T18:45:30.165.354.425 (XO: 20.1654965)

 - Block number: 3 starting at 2017-11-14T18:45:53.733.337.750 (XO: 43.7334801, t=35.6°C)  Rank: 00009
   TM values: Max =    24  Min =   -34  Ave =    -4.59  Std =     7.54
   Min TM value:-34 at an offset of 161.251 usec -> 2017-11-14T18:45:53.733.499.000 (XO: 43.7336414)
   Max TM value:24 at an offset of 163.501 usec -> 2017-11-14T18:45:53.733.501.250 (XO: 43.7336436)
   Max dTM/dt [1/us]::4.22e+02 at an offset of 104.625 usec -> 2017-11-14T18:45:53.733.442.375 (XO: 43.7335847)
   Min dTM/dt [1/us]::-4.42e+02 at an offset of 161.226 usec -> 2017-11-14T18:45:53.733.498.975 (XO: 43.7336413)
