-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171114_174435_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 13.740386 to 30.468544
-each separately from 13s to 30s and from 468255us to 740675us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.802571 in 1 packets - #/No: 18/18
      8.924720 in 1 packets - #/No: 36/18
MIU packet time from 2017-11-14T17:44:51.740.386.000 to 2017-11-14T17:45:08.468.544.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.802571 in 1 packets - #/No: 18/18
      8.924720 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.4359968 to 31.4359347
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      7.9999708 in 1 packets - #/No: 18/18
      8.9999671 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 54 packets (all packets)
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171114_174435_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.1763816 to 15.1766706
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.8025430  -- OK for SL --
   MIU time from 2017-11-14T17:44:51.740.386.000 to 2017-11-14T17:44:51.740.675.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.802571

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      17 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.9792136 to 22.9795026
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.9246870  -- OK for SL --
   MIU time from 2017-11-14T17:44:59.543.246.000 to 2017-11-14T17:44:59.543.535.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.924720

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.9041896 to 31.9044786
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-11-14T17:45:08.468.255.000 to 2017-11-14T17:45:08.468.544.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-11-14T17:44:51.740.245.750
   Max SL corrected packet time: 2017-11-14T17:45:08.468.403.750
   Min SL corrected XO rel time: 13.7403875
   Max SL corrected XO rel time: 30.4685456
   TM values from -34 to 33 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-11-14T17:44:51.740.245.750 (XO: 13.7403875, t=35.6°C)  Rank: 00008
   TM values: Max =    21  Min =   -27  Ave =    -4.42  Std =     5.85
   Min TM value:-27 at an offset of 151.788 usec -> 2017-11-14T17:44:51.740.397.538 (XO: 13.7405393)
   Max TM value:21 at an offset of 158.926 usec -> 2017-11-14T17:44:51.740.404.675 (XO: 13.7405464)
   Max dTM/dt [1/us]::2.93e+02 at an offset of 151.838 usec -> 2017-11-14T17:44:51.740.397.588 (XO: 13.7405393)
   Min dTM/dt [1/us]::-2.68e+02 at an offset of 50.575 usec -> 2017-11-14T17:44:51.740.296.325 (XO: 13.7404380)

 - Block number: 2 starting at 2017-11-14T17:44:59.543.105.750 (XO: 21.5432480, t=35.6°C)  Rank: 00017
   TM values: Max =    33  Min =   -34  Ave =    -4.83  Std =     7.02
   Min TM value:-34 at an offset of 107.200 usec -> 2017-11-14T17:44:59.543.212.950 (XO: 21.5433552)
   Max TM value:33 at an offset of 103.288 usec -> 2017-11-14T17:44:59.543.209.037 (XO: 21.5433512)
   Max dTM/dt [1/us]::2.00e+02 at an offset of 98.175 usec -> 2017-11-14T17:44:59.543.203.925 (XO: 21.5433461)
   Min dTM/dt [1/us]::-1.90e+02 at an offset of 198.588 usec -> 2017-11-14T17:44:59.543.304.338 (XO: 21.5434465)

 - Block number: 3 starting at 2017-11-14T17:45:08.468.114.750 (XO: 30.4682566, t=35.6°C)  Rank: 00005
   TM values: Max =    17  Min =   -22  Ave =    -4.51  Std =     4.36
   Min TM value:-22 at an offset of 41.150 usec -> 2017-11-14T17:45:08.468.155.900 (XO: 30.4682978)
   Max TM value:17 at an offset of 47.713 usec -> 2017-11-14T17:45:08.468.162.462 (XO: 30.4683043)
   Max dTM/dt [1/us]::2.08e+02 at an offset of 59.763 usec -> 2017-11-14T17:45:08.468.174.512 (XO: 30.4683164)
   Min dTM/dt [1/us]::-2.06e+02 at an offset of 113.563 usec -> 2017-11-14T17:45:08.468.228.312 (XO: 30.4683702)
