-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171114_164403_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 8.464532 to 32.637228
-each separately from 8s to 32s and from 279437us to 637228us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.814616 in 1 packets - #/No: 18/18
      8.357213 in 1 packets - #/No: 36/18
MIU packet time from 2017-11-14T16:44:14.464.532.000 to 2017-11-14T16:44:38.637.228.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.814616 in 1 packets - #/No: 18/18
      8.357213 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.4325380 to 33.4324514
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.9999424 in 1 packets - #/No: 18/18
      7.9999710 in 1 packets - #/No: 36/18
Temperature FPGA: 
      04 -> 49.8°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      05 -> 48.5°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171114_164403_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      27 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.8970696 to 9.8973586
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.8145590  -- OK for SL --
   MIU time from 2017-11-14T16:44:14.464.532.000 to 2017-11-14T16:44:14.464.821.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.814616

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.7119176 to 25.7122066
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.3571830  -- OK for SL --
   MIU time from 2017-11-14T16:44:30.279.437.000 to 2017-11-14T16:44:30.279.726.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.357213

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.0693896 to 34.0696786
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-11-14T16:44:38.636.939.000 to 2017-11-14T16:44:38.637.228.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-11-14T16:44:14.464.391.750
   Max SL corrected packet time: 2017-11-14T16:44:38.637.087.750
   Min SL corrected XO rel time: 8.4645333
   Max SL corrected XO rel time: 32.6372295
   TM values from -47 to 36 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-11-14T16:44:14.464.391.750 (XO: 8.4645333, t=35.6°C)  Rank: 00027
   TM values: Max =    36  Min =   -47  Ave =    -4.46  Std =     6.87
   Min TM value:-47 at an offset of 95.838 usec -> 2017-11-14T16:44:14.464.487.587 (XO: 8.4646291)
   Max TM value:36 at an offset of 90.513 usec -> 2017-11-14T16:44:14.464.482.262 (XO: 8.4646238)
   Max dTM/dt [1/us]::1.57e+02 at an offset of 295.051 usec -> 2017-11-14T16:44:14.464.686.801 (XO: 8.4648283)
   Min dTM/dt [1/us]::-1.50e+02 at an offset of 166.001 usec -> 2017-11-14T16:44:14.464.557.750 (XO: 8.4646993)

 - Block number: 2 starting at 2017-11-14T16:44:30.279.296.750 (XO: 24.2794382, t=35.6°C)  Rank: 00007
   TM values: Max =    13  Min =   -26  Ave =    -4.53  Std =     4.36
   Min TM value:-26 at an offset of 33.525 usec -> 2017-11-14T16:44:30.279.330.275 (XO: 24.2794717)
   Max TM value:13 at an offset of 87.113 usec -> 2017-11-14T16:44:30.279.383.862 (XO: 24.2795253)
   Max dTM/dt [1/us]::2.43e+02 at an offset of 274.138 usec -> 2017-11-14T16:44:30.279.570.888 (XO: 24.2797123)
   Min dTM/dt [1/us]::-2.15e+02 at an offset of 186.276 usec -> 2017-11-14T16:44:30.279.483.025 (XO: 24.2796245)

 - Block number: 3 starting at 2017-11-14T16:44:38.636.798.750 (XO: 32.6369405, t=35.6°C)  Rank: 00002
   TM values: Max =    11  Min =   -17  Ave =    -4.59  Std =     3.66
   Min TM value:-17 at an offset of 49.688 usec -> 2017-11-14T16:44:38.636.848.437 (XO: 32.6369902)
   Max TM value:11 at an offset of 144.951 usec -> 2017-11-14T16:44:38.636.943.700 (XO: 32.6370854)
   Max dTM/dt [1/us]::1.85e+02 at an offset of 99.050 usec -> 2017-11-14T16:44:38.636.897.800 (XO: 32.6370395)
   Min dTM/dt [1/us]::-1.68e+02 at an offset of 99.800 usec -> 2017-11-14T16:44:38.636.898.550 (XO: 32.6370403)
