-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171110_105546_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.127766 to 30.350702
-each separately from 0s to 30s and from 127766us to 532179us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.403835 in 1 packets - #/No: 18/18
      14.818234 in 1 packets - #/No: 36/18
MIU packet time from 2017-11-10T10:55:49.127.766.000 to 2017-11-10T10:56:19.350.702.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      15.403835 in 1 packets - #/No: 18/18
      14.818234 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.9396641 to 31.9395459
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      14.9999409 in 2 packets - #/No: 18/18, 36/18
Temperature FPGA: 
      04 -> 49.8°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      05 -> 48.5°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171110_105546_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.0674309 to 2.0677199
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.4037750  -- OK for SL --
   MIU time from 2017-11-10T10:55:49.127.766.000 to 2017-11-10T10:55:49.128.055.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.403835

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.4714949 to 17.4717839
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.8181750  -- OK for SL --
   MIU time from 2017-11-10T10:56:04.531.890.000 to 2017-11-10T10:56:04.532.179.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.818234

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.2899589 to 32.2902479
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-11-10T10:56:19.350.413.000 to 2017-11-10T10:56:19.350.702.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-11-10T10:55:49.127.625.750
   Max SL corrected packet time: 2017-11-10T10:56:19.350.561.750
   Min SL corrected XO rel time: 0.1277673
   Max SL corrected XO rel time: 30.3507034
   TM values from -18 to 12 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-11-10T10:55:49.127.625.750 (XO: 0.1277673, t=35.6°C)  Rank: 00002
   TM values: Max =     8  Min =   -17  Ave =    -4.50  Std =     3.58
   Min TM value:-17 at an offset of 86.150 usec -> 2017-11-10T10:55:49.127.711.900 (XO: 0.1278535)
   Max TM value:8 at an offset of 115.825 usec -> 2017-11-10T10:55:49.127.741.575 (XO: 0.1278831)
   Max dTM/dt [1/us]::9.30e+01 at an offset of 115.838 usec -> 2017-11-10T10:55:49.127.741.587 (XO: 0.1278831)
   Min dTM/dt [1/us]::-9.50e+01 at an offset of 305.989 usec -> 2017-11-10T10:55:49.127.931.738 (XO: 0.1280733)

 - Block number: 2 starting at 2017-11-10T10:56:04.531.749.750 (XO: 15.5318920, t=35.6°C)  Rank: 00003
   TM values: Max =    12  Min =   -18  Ave =    -4.45  Std =     3.82
   Min TM value:-18 at an offset of 134.838 usec -> 2017-11-10T10:56:04.531.884.588 (XO: 15.5320269)
   Max TM value:12 at an offset of 304.714 usec -> 2017-11-10T10:56:04.532.054.463 (XO: 15.5321967)
   Max dTM/dt [1/us]::1.07e+02 at an offset of 276.739 usec -> 2017-11-10T10:56:04.532.026.488 (XO: 15.5321688)
   Min dTM/dt [1/us]::-1.06e+02 at an offset of 249.313 usec -> 2017-11-10T10:56:04.531.999.063 (XO: 15.5321413)

 - Block number: 3 starting at 2017-11-10T10:56:19.350.272.750 (XO: 30.3504144, t=35.6°C)  Rank: 00002
   TM values: Max =     5  Min =   -15  Ave =    -4.51  Std =     2.68
   Min TM value:-15 at an offset of 133.126 usec -> 2017-11-10T10:56:19.350.405.875 (XO: 30.3505475)
   Max TM value:5 at an offset of 106.750 usec -> 2017-11-10T10:56:19.350.379.500 (XO: 30.3505211)
   Max dTM/dt [1/us]::8.10e+01 at an offset of 138.376 usec -> 2017-11-10T10:56:19.350.411.125 (XO: 30.3505527)
   Min dTM/dt [1/us]::-1.05e+02 at an offset of 140.101 usec -> 2017-11-10T10:56:19.350.412.850 (XO: 30.3505545)
