-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171109_150411_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 2.864122 to 41.350145
-each separately from 2s to 41s and from 195909us to 864411us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      23.331498 in 1 packets - #/No: 18/18
      15.153658 in 1 packets - #/No: 36/18
MIU packet time from 2017-11-09T15:04:16.864.122.000 to 2017-11-09T15:04:55.350.145.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      23.331498 in 1 packets - #/No: 18/18
      15.153658 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.9379168 to 42.9377652
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      23.9999067 in 1 packets - #/No: 18/18
      14.9999417 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      04 -> 49.8°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171109_150411_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      11 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.8020374 to 4.8023264
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.3314070  -- OK for SL --
   MIU time from 2017-11-09T15:04:16.864.122.000 to 2017-11-09T15:04:16.864.411.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.331498

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.1337334 to 28.1340224
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.1535990  -- OK for SL --
   MIU time from 2017-11-09T15:04:40.195.909.000 to 2017-11-09T15:04:40.196.198.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.153658

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      10 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.2876214 to 43.2879104
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-11-09T15:04:55.349.856.000 to 2017-11-09T15:04:55.350.145.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-11-09T15:04:16.863.981.750
   Max SL corrected packet time: 2017-11-09T15:04:55.350.004.750
   Min SL corrected XO rel time: 2.8641240
   Max SL corrected XO rel time: 41.3501465
   TM values from -29 to 40 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-11-09T15:04:16.863.981.750 (XO: 2.8641240, t=35.6°C)  Rank: 00011
   TM values: Max =    40  Min =   -23  Ave =    -4.36  Std =     4.31
   Min TM value:-23 at an offset of 104.988 usec -> 2017-11-09T15:04:16.864.086.737 (XO: 2.8642290)
   Max TM value:40 at an offset of 96.688 usec -> 2017-11-09T15:04:16.864.078.437 (XO: 2.8642207)
   Max dTM/dt [1/us]::1.77e+02 at an offset of 96.525 usec -> 2017-11-09T15:04:16.864.078.275 (XO: 2.8642205)
   Min dTM/dt [1/us]::-1.63e+02 at an offset of 23.000 usec -> 2017-11-09T15:04:16.864.004.750 (XO: 2.8641470)

 - Block number: 2 starting at 2017-11-09T15:04:40.195.768.750 (XO: 26.1959107, t=35.6°C)  Rank: 00008
   TM values: Max =    24  Min =   -21  Ave =    -4.29  Std =     3.94
   Min TM value:-21 at an offset of 89.725 usec -> 2017-11-09T15:04:40.195.858.475 (XO: 26.1960004)
   Max TM value:24 at an offset of 83.288 usec -> 2017-11-09T15:04:40.195.852.037 (XO: 26.1959940)
   Max dTM/dt [1/us]::1.40e+02 at an offset of 43.338 usec -> 2017-11-09T15:04:40.195.812.087 (XO: 26.1959540)
   Min dTM/dt [1/us]::-1.29e+02 at an offset of 43.700 usec -> 2017-11-09T15:04:40.195.812.450 (XO: 26.1959544)

 - Block number: 3 starting at 2017-11-09T15:04:55.349.715.750 (XO: 41.3498575, t=35.6°C)  Rank: 00010
   TM values: Max =    19  Min =   -29  Ave =    -4.42  Std =     4.51
   Min TM value:-29 at an offset of 122.113 usec -> 2017-11-09T15:04:55.349.837.862 (XO: 41.3499796)
   Max TM value:19 at an offset of 114.950 usec -> 2017-11-09T15:04:55.349.830.700 (XO: 41.3499725)
   Max dTM/dt [1/us]::1.27e+02 at an offset of 293.176 usec -> 2017-11-09T15:04:55.350.008.926 (XO: 41.3501507)
   Min dTM/dt [1/us]::-1.21e+02 at an offset of 94.063 usec -> 2017-11-09T15:04:55.349.809.812 (XO: 41.3499516)
