-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171103_021052_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 14.342205 to 32.218531
-each separately from 14s to 32s and from 218242us to 342494us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      0.892370 in 1 packets - #/No: 18/18
      16.983089 in 1 packets - #/No: 36/18
MIU packet time from 2017-11-03T02:11:09.342.205.000 to 2017-11-03T02:11:27.218.531.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      0.892370 in 1 packets - #/No: 18/18
      16.983089 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.4356649 to 33.4355954
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      0.9999961 in 1 packets - #/No: 18/18
      16.9999344 in 1 packets - #/No: 36/18
Temperature FPGA: 
      06 -> 47.2°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      05 -> 48.5°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      BF -> 34.0°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171103_021052_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.7778696 to 15.7781586
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 0.8923670  -- OK for SL --
   MIU time from 2017-11-03T02:11:09.342.205.000 to 2017-11-03T02:11:09.342.494.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 0.892370

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.6705256 to 16.6708146
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.9830230  -- OK for SL --
   MIU time from 2017-11-03T02:11:10.234.864.000 to 2017-11-03T02:11:10.235.153.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.983089

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.6538376 to 33.6541266
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-11-03T02:11:27.218.242.000 to 2017-11-03T02:11:27.218.531.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-11-03T02:11:09.342.064.750
   Max SL corrected packet time: 2017-11-03T02:11:27.218.390.750
   Min SL corrected XO rel time: 14.3422060
   Max SL corrected XO rel time: 32.2185321
   TM values from -19 to 8 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-11-03T02:11:09.342.064.750 (XO: 14.3422060, t=34.0°C)  Rank: 00004
   TM values: Max =     8  Min =   -19  Ave =    -4.33  Std =     3.06
   Min TM value:-19 at an offset of 95.300 usec -> 2017-11-03T02:11:09.342.160.050 (XO: 14.3423013)
   Max TM value:8 at an offset of 92.950 usec -> 2017-11-03T02:11:09.342.157.700 (XO: 14.3422990)
   Max dTM/dt [1/us]::1.48e+02 at an offset of 154.551 usec -> 2017-11-03T02:11:09.342.219.300 (XO: 14.3423606)
   Min dTM/dt [1/us]::-1.60e+02 at an offset of 154.788 usec -> 2017-11-03T02:11:09.342.219.538 (XO: 14.3423608)

 - Block number: 2 starting at 2017-11-03T02:11:10.234.723.750 (XO: 15.2348655, t=34.0°C)  Rank: 00002
   TM values: Max =     7  Min =   -15  Ave =    -4.39  Std =     3.05
   Min TM value:-15 at an offset of 72.063 usec -> 2017-11-03T02:11:10.234.795.812 (XO: 15.2349376)
   Max TM value:7 at an offset of 267.239 usec -> 2017-11-03T02:11:10.234.990.988 (XO: 15.2351327)
   Max dTM/dt [1/us]::1.54e+02 at an offset of 267.201 usec -> 2017-11-03T02:11:10.234.990.951 (XO: 15.2351327)
   Min dTM/dt [1/us]::-1.77e+02 at an offset of 267.289 usec -> 2017-11-03T02:11:10.234.991.038 (XO: 15.2351328)

 - Block number: 3 starting at 2017-11-03T02:11:27.218.101.750 (XO: 32.2182431, t=34.0°C)  Rank: 00002
   TM values: Max =     5  Min =   -14  Ave =    -4.44  Std =     2.52
   Min TM value:-14 at an offset of 79.875 usec -> 2017-11-03T02:11:27.218.181.625 (XO: 32.2183229)
   Max TM value:5 at an offset of 111.950 usec -> 2017-11-03T02:11:27.218.213.700 (XO: 32.2183550)
   Max dTM/dt [1/us]::1.40e+02 at an offset of 79.925 usec -> 2017-11-03T02:11:27.218.181.675 (XO: 32.2183230)
   Min dTM/dt [1/us]::-1.32e+02 at an offset of 91.688 usec -> 2017-11-03T02:11:27.218.193.437 (XO: 32.2183347)
