-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171101_031604_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 12.961297 to 31.012826
-each separately from 12s to 31s and from 12537us to 961586us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      5.060084 in 1 packets - #/No: 18/18
      12.990578 in 1 packets - #/No: 36/18
MIU packet time from 2017-11-01T03:16:19.961.297.000 to 2017-11-01T03:16:38.012.826.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      5.060084 in 1 packets - #/No: 18/18
      12.990578 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.4417885 to 32.4417142
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      5.9999765 in 1 packets - #/No: 18/18
      12.9999492 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 54 packets (all packets)
Temperature XO: 
      BF -> 34.0°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      C0 -> 35.6°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171101_031604_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.4030839 to 14.4033729
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.0600630  -- OK for SL --
   MIU time from 2017-11-01T03:16:19.961.297.000 to 2017-11-01T03:16:19.961.586.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.060084

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.4634359 to 19.4637249
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.9905270  -- OK for SL --
   MIU time from 2017-11-01T03:16:25.021.670.000 to 2017-11-01T03:16:25.021.959.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.990578

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.4542519 to 32.4545409
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-11-01T03:16:38.012.537.000 to 2017-11-01T03:16:38.012.826.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-11-01T03:16:19.961.156.750
   Max SL corrected packet time: 2017-11-01T03:16:38.012.685.750
   Min SL corrected XO rel time: 12.9612991
   Max SL corrected XO rel time: 31.0128268
   TM values from -23 to 14 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-11-01T03:16:19.961.156.750 (XO: 12.9612991, t=34.0°C)  Rank: 00006
   TM values: Max =    13  Min =   -23  Ave =    -4.44  Std =     6.45
   Min TM value:-23 at an offset of 97.450 usec -> 2017-11-01T03:16:19.961.254.200 (XO: 12.9613966)
   Max TM value:13 at an offset of 116.938 usec -> 2017-11-01T03:16:19.961.273.687 (XO: 12.9614161)
   Max dTM/dt [1/us]::2.92e+02 at an offset of 97.488 usec -> 2017-11-01T03:16:19.961.254.237 (XO: 12.9613966)
   Min dTM/dt [1/us]::-2.87e+02 at an offset of 96.888 usec -> 2017-11-01T03:16:19.961.253.637 (XO: 12.9613960)

 - Block number: 2 starting at 2017-11-01T03:16:25.021.529.750 (XO: 18.0216710, t=34.0°C)  Rank: 00007
   TM values: Max =    14  Min =   -22  Ave =    -4.48  Std =     5.57
   Min TM value:-22 at an offset of 119.163 usec -> 2017-11-01T03:16:25.021.648.912 (XO: 18.0217901)
   Max TM value:14 at an offset of 119.763 usec -> 2017-11-01T03:16:25.021.649.512 (XO: 18.0217907)
   Max dTM/dt [1/us]::2.83e+02 at an offset of 138.438 usec -> 2017-11-01T03:16:25.021.668.188 (XO: 18.0218094)
   Min dTM/dt [1/us]::-2.74e+02 at an offset of 156.226 usec -> 2017-11-01T03:16:25.021.685.975 (XO: 18.0218272)

 - Block number: 3 starting at 2017-11-01T03:16:38.012.396.750 (XO: 31.0125377, t=34.0°C)  Rank: 00007
   TM values: Max =    14  Min =   -23  Ave =    -4.48  Std =     6.39
   Min TM value:-23 at an offset of 162.901 usec -> 2017-11-01T03:16:38.012.559.650 (XO: 31.0127006)
   Max TM value:14 at an offset of 104.600 usec -> 2017-11-01T03:16:38.012.501.350 (XO: 31.0126423)
   Max dTM/dt [1/us]::2.96e+02 at an offset of 133.676 usec -> 2017-11-01T03:16:38.012.530.425 (XO: 31.0126714)
   Min dTM/dt [1/us]::-3.10e+02 at an offset of 146.863 usec -> 2017-11-01T03:16:38.012.543.613 (XO: 31.0126846)
