-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171031_110732_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 4.975071 to 34.120161
-each separately from 4s to 34s and from 119872us to 975360us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      11.957421 in 1 packets - #/No: 18/18
      17.186802 in 1 packets - #/No: 36/18
MIU packet time from 2017-10-31T11:07:39.975.071.000 to 2017-10-31T11:08:09.120.161.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      11.957421 in 1 packets - #/No: 18/18
      17.186802 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.4396964 to 35.4395817
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      11.9999540 in 1 packets - #/No: 18/18
      17.9999313 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      06 -> 47.2°C in 18 packets - #/No: 37/1, 38/2, 39/3, 40/4, 41/5, 42/6, 43/7, 44/8, 45/9, 46/10, 47/11, 48/12, 49/13, 50/14, 51/15, 52/16, 53/17, 54/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171031_110732_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      3 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.4147660 to 6.4150550
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.9573750  -- OK for SL --
   MIU time from 2017-10-31T11:07:39.975.071.000 to 2017-10-31T11:07:39.975.360.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.957421

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      2 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.3724300 to 18.3727190
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.1867350  -- OK for SL --
   MIU time from 2017-10-31T11:07:51.932.781.000 to 2017-10-31T11:07:51.933.070.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.186802

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.5594540 to 35.5597430
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-10-31T11:08:09.119.872.000 to 2017-10-31T11:08:09.120.161.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-10-31T11:07:39.974.930.750
   Max SL corrected packet time: 2017-10-31T11:08:09.120.020.750
   Min SL corrected XO rel time: 4.9750733
   Max SL corrected XO rel time: 34.1201618
   TM values from -20 to 11 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-10-31T11:07:39.974.930.750 (XO: 4.9750733, t=35.6°C)  Rank: 00003
   TM values: Max =    10  Min =   -19  Ave =    -4.54  Std =     4.63
   Min TM value:-19 at an offset of 96.388 usec -> 2017-10-31T11:07:39.975.027.137 (XO: 4.9751697)
   Max TM value:10 at an offset of 90.975 usec -> 2017-10-31T11:07:39.975.021.725 (XO: 4.9751643)
   Max dTM/dt [1/us]::1.12e+02 at an offset of 102.725 usec -> 2017-10-31T11:07:39.975.033.475 (XO: 4.9751760)
   Min dTM/dt [1/us]::-1.11e+02 at an offset of 120.838 usec -> 2017-10-31T11:07:39.975.051.587 (XO: 4.9751941)

 - Block number: 2 starting at 2017-10-31T11:07:51.932.640.750 (XO: 16.9327831, t=35.6°C)  Rank: 00002
   TM values: Max =     9  Min =   -17  Ave =    -4.49  Std =     4.02
   Min TM value:-17 at an offset of 101.650 usec -> 2017-10-31T11:07:51.932.742.400 (XO: 16.9328848)
   Max TM value:9 at an offset of 186.263 usec -> 2017-10-31T11:07:51.932.827.013 (XO: 16.9329694)
   Max dTM/dt [1/us]::1.01e+02 at an offset of 228.788 usec -> 2017-10-31T11:07:51.932.869.538 (XO: 16.9330119)
   Min dTM/dt [1/us]::-1.02e+02 at an offset of 101.325 usec -> 2017-10-31T11:07:51.932.742.075 (XO: 16.9328845)

 - Block number: 3 starting at 2017-10-31T11:08:09.119.731.750 (XO: 34.1198728, t=35.6°C)  Rank: 00004
   TM values: Max =    11  Min =   -20  Ave =    -4.54  Std =     5.23
   Min TM value:-20 at an offset of 124.800 usec -> 2017-10-31T11:08:09.119.856.550 (XO: 34.1199976)
   Max TM value:11 at an offset of 128.513 usec -> 2017-10-31T11:08:09.119.860.263 (XO: 34.1200013)
   Max dTM/dt [1/us]::1.32e+02 at an offset of 169.451 usec -> 2017-10-31T11:08:09.119.901.200 (XO: 34.1200422)
   Min dTM/dt [1/us]::-1.24e+02 at an offset of 142.501 usec -> 2017-10-31T11:08:09.119.874.250 (XO: 34.1200153)
