-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171031_070524_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 10.376479 to 36.347987
-each separately from 10s to 36s and from 290617us to 376768us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.913849 in 1 packets - #/No: 18/18
      19.056792 in 1 packets - #/No: 36/18
MIU packet time from 2017-10-31T07:05:37.376.479.000 to 2017-10-31T07:06:03.347.987.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.913849 in 1 packets - #/No: 18/18
      19.056792 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.4434008 to 37.4433009
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      6.9999730 in 1 packets - #/No: 18/18
      18.9999271 in 1 packets - #/No: 36/18
Temperature FPGA: 
      03 -> 51.1°C in 18 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18
      04 -> 49.8°C in 36 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18, ...
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171031_070524_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      6 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.8198793 to 11.8201683
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.9138230  -- OK for SL --
   MIU time from 2017-10-31T07:05:37.376.479.000 to 2017-10-31T07:05:37.376.768.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.913849

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.7339913 to 18.7342803
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.0567190  -- OK for SL --
   MIU time from 2017-10-31T07:05:44.290.617.000 to 2017-10-31T07:05:44.290.906.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.056792

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      5 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.7909993 to 37.7912883
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-10-31T07:06:03.347.698.000 to 2017-10-31T07:06:03.347.987.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-10-31T07:05:37.376.338.750
   Max SL corrected packet time: 2017-10-31T07:06:03.347.846.750
   Min SL corrected XO rel time: 10.3764800
   Max SL corrected XO rel time: 36.3479888
   TM values from -27 to 17 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-10-31T07:05:37.376.338.750 (XO: 10.3764800, t=35.6°C)  Rank: 00006
   TM values: Max =    16  Min =   -25  Ave =    -4.41  Std =     6.23
   Min TM value:-25 at an offset of 151.126 usec -> 2017-10-31T07:05:37.376.489.875 (XO: 10.3766311)
   Max TM value:16 at an offset of 151.876 usec -> 2017-10-31T07:05:37.376.490.625 (XO: 10.3766318)
   Max dTM/dt [1/us]::1.93e+02 at an offset of 124.525 usec -> 2017-10-31T07:05:37.376.463.275 (XO: 10.3766045)
   Min dTM/dt [1/us]::-1.86e+02 at an offset of 225.013 usec -> 2017-10-31T07:05:37.376.563.763 (XO: 10.3767050)

 - Block number: 2 starting at 2017-10-31T07:05:44.290.476.750 (XO: 17.2906186, t=35.6°C)  Rank: 00007
   TM values: Max =    17  Min =   -27  Ave =    -4.49  Std =     6.20
   Min TM value:-27 at an offset of 186.163 usec -> 2017-10-31T07:05:44.290.662.913 (XO: 17.2908048)
   Max TM value:17 at an offset of 153.101 usec -> 2017-10-31T07:05:44.290.629.850 (XO: 17.2907717)
   Max dTM/dt [1/us]::2.18e+02 at an offset of 122.175 usec -> 2017-10-31T07:05:44.290.598.925 (XO: 17.2907408)
   Min dTM/dt [1/us]::-2.13e+02 at an offset of 153.151 usec -> 2017-10-31T07:05:44.290.629.900 (XO: 17.2907718)

 - Block number: 3 starting at 2017-10-31T07:06:03.347.557.750 (XO: 36.3476998, t=35.6°C)  Rank: 00005
   TM values: Max =    12  Min =   -20  Ave =    -4.42  Std =     5.01
   Min TM value:-20 at an offset of 164.326 usec -> 2017-10-31T07:06:03.347.722.075 (XO: 36.3478641)
   Max TM value:12 at an offset of 265.589 usec -> 2017-10-31T07:06:03.347.823.338 (XO: 36.3479653)
   Max dTM/dt [1/us]::1.44e+02 at an offset of 134.463 usec -> 2017-10-31T07:06:03.347.692.213 (XO: 36.3478342)
   Min dTM/dt [1/us]::-1.45e+02 at an offset of 304.464 usec -> 2017-10-31T07:06:03.347.862.213 (XO: 36.3480042)
