-- OK --
-----------------> BLESK ver.2015-09-12
-- File 20171030_175828_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Packet no from 1 to 18
   Differences of packet numbers in neigboring packets: 
      1 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      -17 in 2 packets - #/No: 18/18, 36/18
Total number of packets: 
      18 in 54 packets (all packets)
Hz and MHz counters from 0.010887 to 32.557502
-each separately from 0s to 32s and from 10887us to 857710us.
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      23.846245 in 1 packets - #/No: 18/18
      8.699503 in 1 packets - #/No: 36/18
MIU packet time from 2017-10-30T17:58:31.010.887.000 to 2017-10-30T17:59:03.557.502.000
   Differences in neigboring packets: 
      0.000017 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      23.846245 in 1 packets - #/No: 18/18
      8.699503 in 1 packets - #/No: 36/18
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.4357421 to 33.4356250
   Differences in neigboring packets: 
      0.0000000 in 51 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 19/1, ...
      22.9999160 in 1 packets - #/No: 18/18
      8.9999669 in 1 packets - #/No: 36/18
Temperature FPGA: 
      05 -> 48.5°C in 36 packets - #/No: 1/1, 2/2, 3/3, 4/4, 5/5, 6/6, 7/7, 8/8, 9/9, 10/10, 11/11, 12/12, 13/13, 14/14, 15/15, 16/16, 17/17, 18/18, ...
      04 -> 49.8°C in 18 packets - #/No: 19/1, 20/2, 21/3, 22/4, 23/5, 24/6, 25/7, 26/8, 27/9, 28/10, 29/11, 30/12, 31/13, 32/14, 33/15, 34/16, 35/17, 36/18
Temperature XO: 
      C0 -> 35.6°C in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0°C in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0°C in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
Zero: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20171030_175828_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      4 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.4466298 to 1.4469188
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.8461590  -- OK for SL --
   MIU time from 2017-10-30T17:58:31.010.887.000 to 2017-10-30T17:58:31.011.176.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.846245

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      7 in 18 packets (all packets)  -- OK --
   10 MHz counter from 25.2930778 to 25.2933668
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.6994710  -- OK for SL --
   MIU time from 2017-10-30T17:58:54.857.421.000 to 2017-10-30T17:58:54.857.710.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.699503

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      8 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.9928378 to 33.9931268
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2017-10-30T17:59:03.557.213.000 to 2017-10-30T17:59:03.557.502.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   Min SL corrected packet time: 2017-10-30T17:58:31.010.746.750
   Max SL corrected packet time: 2017-10-30T17:59:03.557.361.750
   Min SL corrected XO rel time: 0.0108877
   Max SL corrected XO rel time: 32.5575039
   TM values from -45 to 31 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2017-10-30T17:58:31.010.746.750 (XO: 0.0108877, t=35.6°C)  Rank: 00004
   TM values: Max =    12  Min =   -21  Ave =    -4.42  Std =     4.49
   Min TM value:-21 at an offset of 122.263 usec -> 2017-10-30T17:58:31.010.869.012 (XO: 0.0110100)
   Max TM value:12 at an offset of 130.475 usec -> 2017-10-30T17:58:31.010.877.225 (XO: 0.0110182)
   Max dTM/dt [1/us]::2.60e+02 at an offset of 132.075 usec -> 2017-10-30T17:58:31.010.878.825 (XO: 0.0110198)
   Min dTM/dt [1/us]::-2.58e+02 at an offset of 131.188 usec -> 2017-10-30T17:58:31.010.877.937 (XO: 0.0110189)

 - Block number: 2 starting at 2017-10-30T17:58:54.857.280.750 (XO: 23.8574230, t=35.6°C)  Rank: 00007
   TM values: Max =    31  Min =   -45  Ave =    -4.44  Std =    15.89
   Min TM value:-45 at an offset of 222.088 usec -> 2017-10-30T17:58:54.857.502.838 (XO: 23.8576450)
   Max TM value:31 at an offset of 113.825 usec -> 2017-10-30T17:58:54.857.394.575 (XO: 23.8575368)
   Max dTM/dt [1/us]::6.44e+02 at an offset of 114.613 usec -> 2017-10-30T17:58:54.857.395.362 (XO: 23.8575376)
   Min dTM/dt [1/us]::-6.43e+02 at an offset of 113.863 usec -> 2017-10-30T17:58:54.857.394.612 (XO: 23.8575368)

 - Block number: 3 starting at 2017-10-30T17:59:03.557.072.750 (XO: 32.5572149, t=35.6°C)  Rank: 00008
   TM values: Max =    25  Min =   -33  Ave =    -4.49  Std =    12.71
   Min TM value:-33 at an offset of 232.176 usec -> 2017-10-30T17:59:03.557.304.925 (XO: 32.5574470)
   Max TM value:25 at an offset of 123.513 usec -> 2017-10-30T17:59:03.557.196.262 (XO: 32.5573384)
   Max dTM/dt [1/us]::4.77e+02 at an offset of 239.926 usec -> 2017-10-30T17:59:03.557.312.675 (XO: 32.5574548)
   Min dTM/dt [1/us]::-4.66e+02 at an offset of 131.388 usec -> 2017-10-30T17:59:03.557.204.137 (XO: 32.5573462)
